Hybrid radiofrequency and cryoablation device: preliminary results in an animal model.
OBJECTIVE
To determine whether the simultaneous application of combined bipolar radiofrequency (RF) ablation and cryoablation in a hybrid system produces larger ablation zones than RF or cryoablation alone.


METHODS
Multiple 15-minute ablations were performed in ex vivo bovine liver (n = 167) with a hybrid applicator system with RF ablation alone (0.3-0.7 A), cryoablation alone (3,500 psi, two freeze/thaw cycles), and combined RF/cryoablation (0.4-0.7 A, 1,000-3,500 psi) with use of a novel applicator consisting of two 2.5-cm active bipolar RF poles located on the same 18-gauge needle separated by two embedded cryoablation nozzles. Resultant coagulation diameters were compared with use of analysis of variance for more than three groups or Student t tests for two groups. Confirmation of the optimal parameters of combination RF/cryoablation was performed by reassessing a range of argon pressure (1,000-3,500 psi) and RF current (0.4-0.7 A) in in vivo porcine liver (n = 36). Arrays of two to four RF/cryoablation applicators were also assessed in ex vivo (n = 54) and in vivo (n = 12) liver.


RESULTS
In ex vivo liver, simultaneous RF/cryoablation (0.6 A, 3,000 psi) produced 3.6 cm +/- 0.4 of short-axis coagulation. This was significantly larger than that achieved with optimal RF alone or cryoablation alone (1.5 cm +/- 0.3 and 1.6 cm +/- 0.3, respectively; F = 95; P < .01). The coagulation diameter with simultaneous combination RF/cryoablation was related in parabolic fashion to argon pressure and current with a multivariate r(2) of 0.68. For in vivo liver, optimal combination RF/cryoablation achieved 3.3 cm +/- 0.2 of coagulation, which was significantly larger than that achieved with RF alone (1.1 cm +/- 0.1; P < .01) or cryoablation alone (1.1 cm +/- 0.1 and 1.3 cm +/- 0.1; F = 203; P < .01). The greatest contiguous coagulation was achieved with multiple-applicator arrays. For ex vivo liver, short-axis coagulation measured 5.3 cm +/- 0.1, 6.4 cm +/- 0.1, and 7.6 cm +/- 0.1 for two-, three-, and four-applicator arrays, respectively. For in vivo liver, two-, three-, and four-applicator arrays produced 5.1 cm +/- 0.2, 5.8 cm +/- 0.5, and 7.0 cm +/- 0.5 of confluent coagulation, respectively.


CONCLUSIONS
Simultaneous combination RF and cryoablation with use of a novel applicator design yielded significantly larger zones of coagulation than either modality alone. The large ablation diameters achieved warrant further investigation of the device.
Hines-Peralta
Andrew
A
Department of Radiology, Beth Israel Deaconess Medical Center, Harvard Medical School, 1 Deaconess Road, WCC 308B, Boston, Massachusetts 02215, USA.
Hollander
C Yehuda
CY
Solazzo
Stephanie
S
Horkan
Clare
C
Liu
Zheng-Jun
ZJ
Goldberg
S Nahum
SN
D000704:Analysis of Variance; D000818:Animals; D017115:Catheter Ablation / Q000295:instrumentation*; D002417:Cattle; D003452:Cryosurgery / Q000295:instrumentation*; D004867:Equipment Design; D066298:In Vitro Techniques; D008099:Liver / Q000473:pathology*; D012044:Regression Analysis; D013552:Swine