Microfluidic biosensing systems. Part I. Development and optimisation of enzymatic chemiluminescent micro-biosensors based on silicon microchips.
Chemiluminescent (CL) enzyme-based flow-through microchip biosensors (micro-biosensors) for detection of glucose and ethanol were developed for the purpose of monitoring real-time production and release of glucose and ethanol from microchip immobilised yeast cells. Part I of this study focuses on the development and optimisation of the micro-biosensors in a microfluidic sequential injection analysis (microSIA) system. Glucose oxidase (GOX) or alcohol oxidase (AOX) was co-immobilised with horseradish peroxidase (HRP) on porous silicon flow through microchips. The hydrogen peroxide produced from oxidation of the corresponding analyte (glucose or ethanol) took part in the chemiluminescent (CL) oxidation of luminol catalysed by HRP enhanced by addition of p-iodophenol (PIP). All steps in the microSIA system, including control of syringe pump, multiposition valve (MPV) and data readout, were computer controlled. The influence of flow rate and luminol- and PIP concentration were investigated using a 2(3)-factor experiment using the GOX-HRP sensor. It was found that all estimated single factors and the highest order of interaction were significant. The optimum was found at 250 microM luminol and 150 microM PIP at a flow rate of 18 microl min(-1), the latter as a compromise between signal intensity and analysis time. Using the optimised system settings one sample was processed within 5 min. Two different immobilisation chemistries were investigated for both micro-biosensors based on 3-aminopropyltriethoxsilane (APTS)- or polyethylenimine (PEI) functionalisation followed by glutaraldehyde (GA) activation. GOX-HRP micro-biosensors responded linear in a log-log format within the range 10-1000 microM glucose. Both had an operational stability of at least 8 days, but the PEI-GOX-HRP sensor was more sensitive. The AOX-HRP micro-biosensors responded linear (log-log) in the range between 1 and 10 mM ethanol, but the PEI-AOX-HRP sensor was in general more sensitive. Both sensors had an operational stability of at least 8 h, but with a half-life of 2-3 days.
Davidsson
Richard
R
Department of Analytical Chemistry, Lund University, P.O. Box 124, SE-22200 Lund, Sweden.
Genin
Frédéric
F
Bengtsson
Martin
M
Laurell
Thomas
T
Emnéus
Jenny
J
D015374:Biosensing Techniques / Q000295:instrumentation* / Q000379:methods; D004867:Equipment Design; D007124:Immunoenzyme Techniques / Q000295:instrumentation* / Q000379:methods; D008163:Luminescent Measurements; D044085:Microfluidics / Q000295:instrumentation* / Q000379:methods; D012825:Silicon / Q000737:chemistry*
D012825:Silicon