G-scan--mobile multiview 3-D measuring system for the analysis of the face.
The development of optical 3-D measuring techniques and their use in industrial quality assurance, in design, and for rapid prototyping has experienced strong growth. A large number of optical 3-D measuring methods and systems are on the market in dentistry. CAD/CAM production has become firmly established in dental medicine, not least due to the systematic introduction of the Cerec technique and the digiDent method. The scanners on which these technologies are based are designed for a relatively small measuring area. To be able to measure and three-dimensionally assess the face--and the numerous changes in the face/forehead/neck region--it was necessary to design and develop a self-calibrating measuring system with gray code for clinical use: the G-Scan measuring system. Objects up to a size of 500 x 500 x 400 mm can be acquired three-dimensionally with it, with a measuring inaccuracy of 10 to 70 microm in a typical measuring time of 15 s. The present article describes the measuring principle, the system parameters, and the features of the new measuring system, and illustrates the measuring results on 3-D displays of the face in static occlusion and in functional occlusion positions.
Kopp
S
S
Clinic of Orthodontics, Friedrich Schiller University, Jena, Germany. stefan.kopp@med.uni-jena.de
Kühmstedt
P
P
Notni
G
G
Geller
R
R
D016302:Centric Relation; D002508:Cephalometry / Q000379:methods*; D017076:Computer-Aided Design; D005145:Face / Q000033:anatomy & histology*; D006801:Humans; D021621:Imaging, Three-Dimensional / Q000295:instrumentation*; D008027:Light; D023861:Photography, Dental; D015203:Reproducibility of Results; D012984:Software