Mobile, outdoor continuous-flow isotope-ratio mass spectrometer system for automated high-frequency 13C- and 18O-CO2 analysis for Keeling plot applications.
A continuous-flow isotope-ratio mass spectrometer (CF-IRMS, custom-made GasBenchII and Delta(plus)Advantage, ThermoFinnigan) was installed on a grassland site and interfaced with a closed-path infrared gas analyser (IRGA). The CF-IRMS and IRGA were housed in an air-conditioned travel van. Air was sampled at 1.5 m above the 0.07-m tall grassland canopy, drawn through a 17-m long PTFE tube at a rate of 0.25 L s(-1), and fed to the IRGA and CF-IRMS in series. The IRMS was interfaced with the IRGA via a stainless steel capillary inserted 0.5 m into the sample air outlet tube of the IRGA (forming an open split), a gas-tight pump, and a sample loop attached to the eight-port Valco valve of the continuous-flow interface. Air was pumped through the 0.25-mL sample loop at 10 mL s(-1) (a flushing frequency of 40 Hz). Air samples were analysed at intervals of approx. 2.8 min. Whole system precision was tested in the field using air mixed from pure CO2 and CO2-free air by means of mass flow controllers. The standard deviation of repeated single measurements was 0.21-0.07 per thousand for delta13C and 0.34-0.14 per thousand for delta18O of CO2 in air with mixing ratios ranging between 200-800 micromol mol(-1). The CO2 peak area measured by the IRMS was proportional to the CO2 mixing ratio (r2 = 1.00), allowing estimation of sample air CO2 mixing ratio from IRMS data. A 1-day long measurement cycle of CO2, delta13C and delta18O of air sampled above the grassland canopy was used to test the system for Keeling plot applications. Delta18O exhibited a clear diurnal cycle (4 per thousand range), but short-term (1-h interval) variability was small (average SD 0.38 per thousand). Yet, the correlation between delta18O and CO2 mixing ratio was relatively weak, and this was true for both the whole data set and 1-h subsets. Conversely, the delta13C of all 541 samples measured during the 25.2-h interval fitted well the Keeling regression (r2 = 0.99), yielding an intercept of -27.40 per thousand (+/-0.07 per thousand SE). Useful Keeling regressions (r2 > 0.9, average r2 = 0.96) also resulted from data collected over 1-h intervals of the 12-h long twilight and dark period. These indicated that 13C content of ecosystem respiration was approx. constant near -27.6 per thousand. The precision of the present system is similar to that of current techniques used in ecosystem studies which employ flask sampling and a laboratory-based CF-IRMS. Sampling (and measurement) frequency is greatly increased relative to systems based on flask sampling, and sampling time (0.025 s per sample) is decreased. These features increase the probability for sampling the entire CO2 range which occurs in a given time window. The system obviates sample storage problems, greatly minimises handling needs, and allows extended campaigns of high frequency sampling and analysis with minimal attendance.
Schnyder
Hans
H
Lehrstuhl für Grünlandlehre, Technische Universität München, D-85350 Freising-Weihenstephan, Germany. schnyder@wzw.tum.de
Schäufele
Rudi
R
Wenzel
Richard
R
D001272:Atmosphere / Q000737:chemistry*; D001331:Automation / Q000295:instrumentation / Q000379:methods; D002245:Carbon Dioxide / Q000032:analysis* / Q000737:chemistry; D002247:Carbon Isotopes; D017753:Ecosystem; D013058:Mass Spectrometry / Q000295:instrumentation* / Q000379:methods*; D010103:Oxygen Isotopes; D012107:Research Design; D012680:Sensitivity and Specificity; D014197:Trees / Q000378:metabolism
D002247:Carbon Isotopes; D010103:Oxygen Isotopes; D002245:Carbon Dioxide