title: [Comparison of radiation dose and image quality of Siremobil-IsoC(3D) with a 16-slice spiral CT for diagnosis and intervention in the human pelvic bone].
abstract: OBJECTIVE
  To compare the image quality of 16-slice computed tomography with the image quality of Siremobil-IsoC (3D) of the pelvic region and to measure simultaneously the radiation dose before and after implantation of a sacroiliac screw (SI-screw)
  
  
  METHODS
  The pelvic region of 8 human cadavers was examined in the Siremobil-IsoC (3D) at five different levels. We used a standard protocol for the 16-slice CT of the complete pelvic region before and after insertion of a pelvic screw, followed by stepwise reduction of the tube current to find the tube current that equalizes the image quality of both modalities. We controlled the image quality by judging important structures such as neuroforamen, nerves, sacroiliacal joint space, intervertebral space, osteophytes, iliopsoas muscle, acetabular surface, fovea centralis, hip joint and os pubis. The image quality was judged by three radiologists and three trauma surgeons using a ranking from 1 to 5. The dose was measured with an endorectally placed NOMEX Dosimeter, to obtain the gonadal dose.
  
  
  RESULTS
  The medium score for all viewers of the Siremobil-IsoC (3D) examinations was between 3 and 4.3. The medium score for all CT-examinations with a tube current of 250 mA was between 1.3 and 2.2. The reduction of tube current down to 80 mA hardly influenced the marks for the analyzed structures. Under 80 mA, bony structures, even after implantation of a SI-screw, were still marked as good, but soft tissue differentiation was getting worse. For the examination of the pelvis, the average dose-length product for the IsoC (3D) was 41.2 mGy x cm. The medium dose-length product for CT was 389 mGy x cm for 250 mA, 125 mGy x cm for 80 mA and 82 mGy x cm for 60 mA.
  
  
  CONCLUSIONS
  The Siremobil-IsoC (3D) is sufficient for therapeutic intraoperative purpose, but the image quality is not sufficient for diagnostic purpose. The higher dose-length product of a CT examination is justifiable because of a better overview, shorter examination time and qualitative superiority. An advantage of the Siremobil-IsoC (3D) is the intraoperative availability with acceptable 3D image quality compared to conventional fluoroscopy.
authors.affiliation: Klinik für Strahlenheilkunde, Charité, Universitätsmedizin Berlin. gero.wieners@charite.de
authors.forename: G
authors.initials: G
authors.lastname: Wieners
authors.forename: M
authors.initials: M
authors.lastname: Pech
authors.forename: A
authors.initials: A
authors.lastname: Beck
authors.forename: B
authors.initials: B
authors.lastname: König
authors.forename: U
authors.initials: U
authors.lastname: Erdmenger
authors.forename: U
authors.initials: U
authors.lastname: Stöckle
authors.forename: P
authors.initials: P
authors.lastname: Wust
authors.forename: R
authors.initials: R
authors.lastname: Felix
authors.forename: R-J
authors.initials: RJ
authors.lastname: Schröder
mesh_terms: D001863:Bone Screws; D002102:Cadaver; D004867:Equipment Design; D006801:Humans; D010384:Pelvic Bones / Q000000981:diagnostic imaging*; D012446:Sacroiliac Joint / Q000000981:diagnostic imaging; D036542:Tomography, Spiral Computed / Q000295:instrumentation* / Q000379:methods*