Reduction of motion artifact in pulse oximetry by smoothed pseudo Wigner-Ville distribution.
BACKGROUND: The pulse oximeter, a medical device capable of measuring blood oxygen saturation (SpO2), has been shown to be a valuable device for monitoring patients in critical conditions. In order to incorporate the technique into a wearable device which can be used in ambulatory settings, the influence of motion artifacts on the estimated SpO2 must be reduced. This study investigates the use of the smoothed psuedo Wigner-Ville distribution (SPWVD) for the reduction of motion artifacts affecting pulse oximetry. METHODS: The SPWVD approach is compared with two techniques currently used in this field, i.e. the weighted moving average (WMA) and the fast Fourier transform (FFT) approaches. SpO2 and pulse rate were estimated from a photoplethysmographic (PPG) signal recorded when subject is in a resting position as well as in the act of performing four types of motions: horizontal and vertical movements of the hand, and bending and pressing motions of the finger. For each condition, 24 sets of PPG signals collected from 6 subjects, each of 30 seconds, were studied with reference to the PPG signal recorded simultaneously from the subject's other hand, which was stationary at all times. RESULTS AND DISCUSSION: The SPWVD approach shows significant improvement (p < 0.05), as compared to traditional approaches, when subjects bend their finger or press their finger against the sensor. In addition, the SPWVD approach also reduces the mean absolute pulse rate error significantly (p < 0.05) from 16.4 bpm and 11.2 bpm for the WMA and FFT approaches, respectively, to 5.62 bpm. CONCLUSION: The results suggested that the SPWVD approach could potentially be used to reduce motion artifact on wearable pulse oximeters.
Yan
Yong-Sheng
YS
Joint Research Center for Biomedical Engineering, The Chinese University of Hong Kong, Shatin, Hong Kong. ytzhang@ee.cuhk.edu.hk.
Poon
Carmen Cy
CC
Zhang
Yuan-Ting
YT
J Clin Monit. 1987 Apr;3(2):135-8
3295125
Anesthesiology. 1992 Jun;76(6):1018-38
1599088
Crit Care Med. 1994 Jun;22(6):981-5
8205831
Anesth Analg. 2002 Oct;95(4):967-72, table of contents
12351278
Anesth Analg. 2002 Jan;94(1 Suppl):S54-61
11900040
IEEE Trans Biomed Eng. 2000 Aug;47(8):1018-26
10943049
J Clin Monit Comput. 2000;16(7):509-22
12580210
Comput Biol Med. 1996 Mar;26(2):143-59
8904288
Appl Opt. 1998 Nov 1;37(31):7437-46
18301578
Br J Anaesth. 1994 Apr;72(4):388-92
8155436
IEEE Trans Biomed Eng. 2001 Jul;48(7):795-805
11442291
IEEE Trans Biomed Eng. 2001 Apr;48(4):452-61
11322533
Anaesth Intensive Care. 1993 Oct;21(5):543-50
8273873