Transpalpebral tonometry: reliability and comparison with Goldmann applanation tonometry and palpation in healthy volunteers.
OBJECTIVE
TGDc-01 is a new, portable, transpalpebral tonometry device. The aim of this study was to evaluate the reproducibility of this method, including intraobserver and interobserver deviations, and to compare the results with those of Goldmann applanation tonometry and palpation of intraocular pressure (IOP).


METHODS
A total of 40 eyes of 20 healthy volunteers were included. Two independent parallel observers (1 and 2) performed three replicate measurements per eye, respectively, both using TGDc01 tonometry, one observer (3) performed three replicate measurements using Goldmann applanation tonometry, and one observer (4) measured the IOP via palpation. Intraindividual deviations within and between both observers using TGDc-01 tonometry and between all three methods were investigated about clinically relevance by comparison of medians and quartiles, statistically significance by pairwise sign tests. Comparisons between observers and methods were based on averaged IOP values of the three individual measurements for each observer and each patient. Intraobserver deviations were analysed by means of Friedman tests.


RESULTS
Observers 1 and 2 showed a statistically significant intraobserver deviation when using TGCc-01 (Friedman p = 0.007 for observer 1 and p = 0.002 for observer 2). There was no statistically significant interobserver deviation between observers 1 and 2 (sign test p = 0.200); however, in 45% of the eyes interobserver deviations were larger than plus or minus 3 mm Hg. The median intraindividual deviation between TGDc-01 and Goldmann (TGDc-01 minus Goldmann) was 0 mm Hg (interquartile range -1; 2 mm Hg; sign test p = 0.522); but deviations were larger than plus or minus 3 mm Hg for 38% of the 40 eyes. Median intraindividual deviation between TGDc-01 and palpation (TGDc-01 minus palpation) was -2 mm Hg (interquartile range -4; 1 mm Hg; sign test p = 0.018), but deviations were larger than plus or minus 3 mmHg for 43% of eyes. Median intraindividual deviation between Goldmann and palpation (palpation minus Goldmann) was 2 mm Hg (interquartile range -2; 4 mm Hg; p = 0.429), but deviations were larger than plus or minus 3 mm Hg in 48% of the eyes.


CONCLUSIONS
Interobserver deviations using TGDc-01 tonometry and intraindividual deviations between TGDc-01 tonometry, Goldmann applanation tonometry, and palpation of IOP were found to be clinically relevant. Thus, according to our results TGDc-01 could not be established as a substitute or diagnostic alternative method for Goldmann applanation tonometry. But as deviations between TGDc01 and Goldmann turned out smaller than between palpation of IOP and Goldmann, TGDc-01 seems to provide a better choice for tonometry in patients, in whom Goldmann applanation tonometry is not possible.
Troost
A
A
Department of Ophthalmology, Johannes Gutenberg University, 55131 Mainz, Germany. Annette.Troost@gmx.de
Yun
S H
SH
Specht
K
K
Krummenauer
F
F
Schwenn
O
O
D000328:Adult; D004867:Equipment Design; D005260:Female; D006801:Humans; D007429:Intraocular Pressure; D008297:Male; D015588:Observer Variation; D010173:Palpation; D015203:Reproducibility of Results; D014065:Tonometry, Ocular / Q000295:instrumentation / Q000379:methods*
Am J Ophthalmol. 1968 Jul;66(1):89-91
5659317
Eye (Lond). 2001 Feb;15(Pt 1):52-8
11318296
Albrecht Von Graefes Arch Klin Exp Ophthalmol. 1976 Jan 16;198(1):39-43
1083154
Graefes Arch Clin Exp Ophthalmol. 2005 Apr;243(4):313-6
15864620
Acta Ophthalmol (Copenh). 1993 Apr;71(2):185-8
8333263
Ophthalmologe. 2005 Jan;102(1):70-6
15322801
Ophthalmologe. 2001 Oct;98(10):944-9
11699315
Lancet. 1986 Feb 8;1(8476):307-10
2868172
Am J Ophthalmol. 1960 May;49:967-78
14443098
Graefes Arch Clin Exp Ophthalmol. 2001 Dec;239(12):889-92
11820692
Comput Methods Programs Biomed. 2000 May;62(1):21-34
10699682
Graefes Arch Clin Exp Ophthalmol. 2005 Jun;243(6):563-9
15614540
Can J Ophthalmol. 1982 Jun;17(3):93-5
7116219
Ophthalmologe. 2005 Feb;102(2):158-62
15290200