The effect of venous pulsation on the forehead pulse oximeter wave form as a possible source of error in Spo2 calculation.
Reflective forehead pulse oximeter sensors have recently been introduced into clinical practice. They reportedly have the advantage of faster response times and immunity to the effects of vasoconstriction. Of concern are reports of signal instability and erroneously low Spo(2) values with some of these new sensors. During a study of the plethysmographic wave forms from various sites (finger, ear, and forehead) it was noted that in some cases the forehead wave form became unexpectedly complex in configuration. The plethysmographic signals from 25 general anesthetic cases were obtained, which revealed the complex forehead wave form during 5 cases. We hypothesized that the complex wave form was attributable to an underlying venous signal. It was determined that the use of a pressure dressing over the sensor resulted in a return of a normal plethysmographic wave form. Further examination of the complex forehead wave form reveal a morphology consistent with a central venous trace with atrial, cuspidal, and venous waves. It is speculated that the presence of the venous signal is the source of the problems reported with the forehead sensors. It is believed that the venous wave form is a result of the method of attachment rather than the use of reflective plethysmographic sensors.
Shelley
Kirk H
KH
Department of Anesthesiology, Yale University, New Haven, Connecticut.
Tamai
Doris
D
Jablonka
Denis
D
Gesquiere
Michael
M
Stout
Robert G
RG
Silverman
David G
DG
D005260:Female; D005546:Forehead; D006801:Humans; D008297:Male; D010092:Oximetry; D010100:Oxygen / Q000097:blood; D010991:Plethysmography; D014680:Veins / Q000502:physiology*
D010100:Oxygen
Cooke J, Scharf J. Improving pulse oximeter performance (abstract). Anesthesiology 2002;96:A593.
Bebout DE, Mannheimer PD. Effects of cold-induced peripheral vasoconstriction on pulse amplitude at various pulse oximeter sensor sites (abstract). Anesthesiology 2002;96:A558.
Whitman RA, Garrison ME, Oestreich TJ, Musumbi MS. Evaluation of a new reflectance forehead sensor in detecting oxygen desaturation in patients undergoing polysomnography (abstract). Anesthesiology 2003;99:A553.
Redford DT, Barker SJ, Lichtenthal PR. Evaluation of 2 forehead reflectance oximeters in intraoperative surgical patients (abstract). Anesth Analg 2004;98(S10):A26.
Mahajan A, Lee E, Callom-Moldovan A. Intraoperative use of forehead reflectance oximetry in pediatric patients (abstract). Anesth Analg 2004;98(S7):A20.
Lawson D, Norley I, Korbon G, et al. Blood flow limits and pulse oximeter signal detection. Anesthesiology 1987;67:599–603.
Dorlas JC, Nijboer JA. Photo-electric plethysmography as a monitoring device in anaesthesia: application and interpretation. Br J Anaesth 1985;57:524–30.
Sami HM, Kleinman BS, Lonchyna VA. Central venous pulsations associated with a falsely low oxygen saturation measured by pulse oximetry. J Clin Monit 1991;7:309–12.
Mark JB. Central venous pressure monitoring: clinical insights beyond the numbers. J Cardiothorac Vasc Anesth 1991;5:163–73.
Shelley K, Dickstein M, Shulman S. The detection of peripheral venous pulsation using the pulse oximeter as a plethysmograph. J Clin Monit 1993;9:283–7.