A testing machine for dental air-turbine handpiece characteristics: free-running speed, stall torque, bearing resistance.
The measurement of performance characteristics of dental air turbine handpieces is of interest with respect to product comparisons, standards specifications and monitoring of bearing longevity in clinical service. Previously, however, bulky and expensive laboratory equipment was required. A portable test machine is described for determining three key characteristics of dental air-turbine handpieces: free-running speed, stall torque and bearing resistance. It relies on a special circuit design for performing a hardware integration of a force signal with respect to rotational position, independent of the rate at which the turbine is allowed to turn during both stall torque and bearing resistance measurements. Free-running speed without the introduction of any imbalance can be readily monitored. From the essential linear relationship between torque and speed, dynamic torque and, hence, power, can then be calculated. In order for these measurements to be performed routinely with the necessary precision of location on the test stage, a detailed procedure for ensuring proper gripping of the handpiece is described. The machine may be used to verify performance claims, standard compliance checks should this be established as appropriate, monitor deterioration with time and usage in the clinical environment and for laboratory investigation of design development.
Darvell
Brain W
BW
Dental Materials Science, Faculty of Dentistry, The Prince Philip Dental Hospital, The University of Hong Kong, Hong Kong. b.w.darvell@hku.hk
Dyson
J E
JE
D000388:Air; D003754:Dental High-Speed Equipment; D004867:Equipment Design; D019544:Equipment Failure Analysis / Q000295:instrumentation*; D017276:Friction; D008422:Materials Testing / Q000295:instrumentation; D012399:Rotation; D019415:Torque; D014159:Transducers