Commissioning and quality assurance for intensity modulated radiotherapy with dynamic multileaf collimator: experience of the Pontificia Universidad Católica de Chile.
The objective of this paper is to present our experience in the commissioning and quality assurance (QA) for intensity modulated radiotherapy (IMRT) using dynamic multileaf collimator (dMLC), sliding window technique. Using Varian equipment solution, the connectivity and operation between all IMRT chain components was checked. Then the following test were done: stability of leaf positioning and leaf speed, sensitivity to treatment interruptions (acceleration and deceleration), evaluation of standard field patterns, stability of dMLC output, segmental dose accuracy check, average leaf transmission, dosimetric leaf separation, effects of lateral disequilibrium between adjacent leaves in dose profiles and multiple carriage field verification. Standard patterns were generated for verification: uniform field, pyramid, hole, wedge, peaks and chair. Weekly QA Protocol include: sweeping gap output, Garden Fence Test (narrow bands, 2 mm wide, of exposure spaced at 2-cm intervals) and segmental dose accuracy check. Monthly QA include: sweeping gap output at multiple gantry and collimator angle, sweeping gap output off-axis, Picket Fence Test (eight consecutive movements of a 5-cm wide rectangular field spaced at 5-cm intervals), stability of leaf speed and leaf motor current test (PWM test). Patient QA procedure consists of an absolute dose measurement for all treatments fields in the treatment condition, analysis of actual leaf position versus planned leaf position (dynalog files) for each treatment field, film relative dose determination for each field, film relative dose determination for the plan (all treatment fields) in two axial planes and patient positioning verification with orthogonal films. The tests performed showed acceptable result. After more than one year of IMRT treatment the routine QA machine checks confirm the precision and stability of the IMRT system.
Venencia
Carlos Daniel
CD
Centro de Cáncer Nuestra Señora de la Esperanza Pontificia, Universidad Católica de Chile, Diagonal Paraguay 319, Santiago, Chile. venencia@med.puc.cl
Besa
Pelayo
P
D002677:Chile; D004867:Equipment Design / Q000592:standards; D019544:Equipment Failure Analysis / Q000592:standards; D007558:Italy; D011785:Quality Assurance, Health Care / Q000379:methods* / Q000592:standards; D011874:Radiometry / Q000295:instrumentation / Q000379:methods* / Q000592:standards; D011879:Radiotherapy Dosage; D011880:Radiotherapy Planning, Computer-Assisted / Q000379:methods* / Q000592:standards; D020266:Radiotherapy, Conformal / Q000295:instrumentation* / Q000379:methods / Q000592:standards