rf enhancement and shielding in MRI caused by conductive implants: dependence on electrical parameters for a tube model.
Radio frequency (rf) eddy-currents induced in implants made of conductive material might cause significant image artifacts in magnetic resonance imaging (MRI) such as shielding of the lumen of vascular stents. rf alteration near metal parts was assessed theoretically in the approximation of alternating current electrodynamics: The implant was modeled as tube with diameter d(o), resistance R, and reactance Y, constituting the secondary winding of a transformer. The transmitter coil of the scanner acted as primary winding and generated the linearly polarized rf field B1,app. Tube axis was assumed parallel to B1,app. The results of the calculations were as follows: Ninety percent of the applied rf-field amplitude is reached in the lumen at a ratio chi=R/Y approximately 2. A rapid drop occurs with the reduction of chi, whereas a further increase of chi causes only a small effect. With chi approximately 1/d(o)(Y approximately d2o,R approximately d(o)), conditions for rf alteration clearly depend on the diameter of the tube. Inside tubes with smaller diameter, rf shielding is less pronounced. rf alteration increases in good approximation with the square root of the strength of the static field B0. The following experiments were carried out: Tubes of similar diameter (d(o) approximately 8 mm) made of material of different conductivity (Cu, Nitinol, carbon fiber reinforced plastic with three different fiber structures) were examined at B0=0.2 and 1.5 T in water phantoms. Tube axis was aligned perpendicular to B0 and spin-echo technique was applied. Local rf enhancement near the outer surface of the metal tubes was detected applying manual reduction of the transmitter amplitude. Shielding inside a carbon fiber tube with d(o) approximately 8 mm and inside a smaller tube with d(o)=3.3 mm was compared. Both tubes showed the same wall structure and thickness (d(w)=0.4 mm). All measurements confirmed the theoretical results. Consequences for the construction of vascular stents are discussed, as well as problems with image artifacts due to rf enhancement near solid conductive implants.
Graf
Hansjörg
H
Section on Experimental Radiology, University Hospital Tübingen, D-72076 Tübingen, Germany.
Steidle
Günter
G
Lauer
Ulrike A
UA
Schick
Fritz
F
D000818:Animals; D016477:Artifacts; D001807:Blood Vessel Prosthesis; D001808:Blood Vessels / Q000473:pathology*; D004553:Electric Conductivity; D004574:Electromagnetic Fields; D004867:Equipment Design; D019544:Equipment Failure Analysis; D006801:Humans; D018810:Magnetic Resonance Angiography / Q000295:instrumentation / Q000379:methods*; D019047:Phantoms, Imaging; D011856:Radiographic Image Enhancement / Q000379:methods*; D011857:Radiographic Image Interpretation, Computer-Assisted / Q000379:methods*; D015607:Stents; D014656:Vascular Surgical Procedures / Q000295:instrumentation