title: Calibration of the NPL secondary standard radionuclide calibrator for the new 10R Schott, type 1+ vials.
abstract: For many years, P6 vials have been used for the distribution of a wide range of diagnostic and therapeutic radioactive solutions. The activity measurements were performed in this geometry and, in time, the UK calibration system for nuclear medicine was based on this container as a standard. However, one major supplier of radiopharmaceuticals has replaced the P6 vial with the 10R Type 1+ Schott vial. As the dimensions of the new vial are different from those of the P6 vial and the responses of radionuclide calibrators are known to be container dependent, the need for re-calibration became apparent. Preliminary measurements made on some typical radionuclide calibrators for (125)I solution indicated a difference in response of about 10% between the two vials. The master ionisation chamber of the NPL secondary standard radionuclide calibrator has been re-calibrated and new calibration factors and volume correction factors for 10R Schott vials have been derived for the relevant medical radionuclides. The standard holder was also modified to accommodate the new larger vial. The complete list of factors and the method used to determine them is presented in this paper. The availability of these new factors will improve the quality of activity measurements in nuclear medicine, as calibration services can now be provided by NPL for the new container. These factors can also be employed for all commercial NPL secondary standard radionuclide calibrators (now known as the NPL-CRC and previously as the 671 or ISOCAL IV).
authors.affiliation: Quality of Life Division, National Physical Laboratory, Ionising Radiation Metrology Consultants Ltd., 152 Broom Road, Teddington, Middlesex TW11 0LW, UK. mike.woods@blueyonder.co.uk
authors.forename: M
authors.initials: M
authors.lastname: Baker