[How safe is magnetic resonance imaging in patients with contraceptive implants?].
When patients with an implanted contraceptive device undergo MRI, it must be ensured that the examination involves no risk to the patient (MR safety) and that the diagnosis is not affected by artifacts or the function of the device compromised (MR compatibility). Two basic types of intrauterine devices can be distinguished: the metal-containing/metal-free intrauterine device (IUD) and the hormone-containing implant, the fully metal-free intrauterine system (IUS), as well as the ESSURE insert made of stainless steel, which has been approved for use in Europe since February 2001. The metal-containing and metal-free IUDs and ESSURE are MRI compatible up to a magnetic field strength of 1.5 T. They do not interact in any relevant way with the external magnetic or high-frequency field and the temperature increase is within the physiologic range. The implants merely produce a local signal void with a shape that depends on their orientation relative to the magnetic field lines. At 3 T, only the metal-free IUD and the IUS are MRI safe in terms of the material used. In contrast, metal-containing IUDs and the ESSURE have not yet been fully evaluated in the 3 T field, which is why they represent a contraindication to MRI. No data are available on the MRI compatibility at 3 T for any of these devices.
Mühler
M
M
Institut für Radiologie, Charité-Universitätsmedizin Berlin, Schumannstrasse 20/21, 10117 Berlin. matthias.muehler@charite.de
Taupitz
M
M
D002056:Burns / Q000209:etiology* / Q000517:prevention & control*; D004868:Equipment Failure; D004869:Equipment Safety / Q000379:methods*; D005260:Female; D006801:Humans; D007434:Intrauterine Devices / Q000009:adverse effects*; D008279:Magnetic Resonance Imaging / Q000009:adverse effects* / Q000295:instrumentation; D011832:Radiation Injuries / Q000209:etiology* / Q000517:prevention & control*; D018570:Risk Assessment; D012307:Risk Factors