MCG measurement in the environment of active magnetic shield.
MCG (Magnetocardiography) measurement by a SQUID gradiometer was attempted with only active magnetic shielding (active shielding). A three-axis-canceling-coil active shielding system, where three 16-10-16 turns-coil sets were put in the orthogonal directions, produces a homogeneous magnetic field in a considerable volume surrounding the center. Fluxgate sensors were used as the reference sensors of the system. The system can reduce environmental magnetic noise at low frequencies of less than a few Hz, at 50 Hz and at 150 Hz. Reducing such disturbances stabilizes biomagnetic measurement conditions for SQUIDs in the absence of magnetically shielded rooms (MSR). After filtering and averaging the measured MCG data by a first-order SQUID gradiometer with only the active shielding during the daytime, the QRS complex and T wave was clearly presented.
Yamazaki
K
K
Takenaka Corp., Japan. yamazaki.keita@takenaka.co.jp
Kato
K
K
Kobayashi
K
K
Igarashi
A
A
Sato
T
T
Haga
A
A
Kasai
N
N
D004560:Electricity; D004562:Electrocardiography / Q000379:methods*; D004574:Electromagnetic Fields; D004777:Environment; D011835:Radiation Protection / Q000379:methods*