Discordance between viral loads determined by Roche COBAS AMPLICOR human immunodeficiency virus type 1 monitor (version 1.5) Standard and ultrasensitive assays caused by freezing patient plasma in centrifuged becton-dickinson vacutainer brand plasma preparation tubes.
The Roche COBAS AMPLICOR human immunodeficiency virus type 1 (HIV-1) Monitor (version 1.5) standard and ultrasensitive viral load assays often gave discordant results, with viral loads from the standard assay exceeding those from the ultrasensitive assay by more than 0.5 log(10) for approximately 20% of specimens received. We began studies to determine the extent, magnitude, and reproducibility of the discordance between the assays and to discover and eliminate the cause of this discordance. Until then, we revised our standard operating procedure to include both standard and ultrasensitive testing on all specimens submitted for viral load determinations. Discordant results usually recurred on retesting. They were most prevalent for specimens with ultrasensitive viral loads of <1,000 and rare for specimens with viral loads of >10,000. Often, standard assay results exceeded those of the ultrasensitive assay by 50- to 100-fold. At higher viral loads, the difference between the standard and ultrasensitive assays persisted, but the percent difference was smaller and rarely caused discordance. The proportion of discordant results was significantly higher in specimens from pediatric patients than in specimens from adults. The ultrasensitive viral load determinations generally agreed with the results of the B-DNA (Bayer) viral load assays. If the plasma was transferred from the centrifuged plasma preparation tubes before freezing, standard and ultrasensitive results were concordant with each other and with values determined on plasma from lavender-topped EDTA tubes.
Salimnia
Hossein
H
Department of Pathology, Detroit, MI 48201, USA. hsalimni@dmc.org
Moore
Ellen C
EC
Crane
Lawrence R
LR
Macarthur
Rodger D
RD
Fairfax
Marilynn R
MR
D000293:Adolescent; D000328:Adult; D021121:Branched DNA Signal Amplification Assay; D002648:Child; D002675:Child, Preschool; D004492:Edetic Acid; D005615:Freezing; D015658:HIV Infections / Q000821:virology; D015497:HIV-1 / Q000235:genetics / Q000302:isolation & purification* / Q000502:physiology; D006801:Humans; D021141:Nucleic Acid Amplification Techniques / Q000379:methods; D010949:Plasma; D012367:RNA, Viral / Q000097:blood*; D011933:Reagent Kits, Diagnostic; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity; D013048:Specimen Handling / Q000295:instrumentation* / Q000379:methods*; D019562:Viral Load
D012367:RNA, Viral; D011933:Reagent Kits, Diagnostic; D004492:Edetic Acid