New concepts and new design of permanent maglev rotary artificial heart blood pumps.
According to tradition, permanent maglev cannot achieve stable equilibrium. The authors have developed, to the contrary, two stable permanent maglev impeller blood pumps. The first pump is an axially driven uni-ventricular assist pump, in which the rotor with impeller is radially supported by two passive magnetic bearings, but has one point contact with the stator axially at standstill. As the pump raises its rotating speed, the increasing hydrodynamic force of fluid acting on the impeller will make the rotor taking off from contacting point and disaffiliate from the stator. Then the rotor becomes fully suspended. The second pump is a radially driven bi-ventricular assist pump, i.e., an impeller total artificial heart. Its rotor with two impellers on both ends is supported by two passive magnetic bearings, which counteract the attractive force between rotor magnets and stator coil iron core. The rotor is affiliated to the stator radially at standstill and becomes levitated during rotation. Therefore, the rotor keeps concentric with stator during rotation but eccentric at standstill, as is confirmed by rotor position detection with Honeywell sensors. It concludes that the permanent maglev needs action of a non-magnetic force to achieve stability but a rotating magnetic levitator with high speed and large inertia can maintain its stability merely with passive magnetic bearings.
Qian
K X
KX
Institute of Biomedical Engineering, Jiangsu University, Xue-fu Road 310, Zhenjiang, Jiangsu 212013, China. swyx@ujs.edu.cn
Zeng
P
P
Ru
W M
WM
Yuan
H Y
HY
D004867:Equipment Design; D019544:Equipment Failure Analysis; D006353:Heart-Assist Devices; D006801:Humans; D008280:Magnetics; D012399:Rotation