Determination of gross alpha, 224Ra, 226Ra, and 228Ra activities in drinking water using a single sample preparation procedure.
The current federal and New Jersey State regulations have greatly increased the number of gross alpha and radium tests for public and private drinking water supplies. The determination of radium isotopes in water generally involves lengthy and complicated processes. In this study, a new approach is presented for the determination of gross alpha, 224Ra, 226Ra, and 228Ra activities in water samples. The method includes a single sample preparation procedure followed by alpha counting and gamma-ray spectroscopy. The sample preparation technique incorporates an EPA-approved co-precipitation methodology for gross alpha determination with a few alterations and improvements. Using 3-L aliquots of sample, spiked with 133Ba tracer, the alpha-emitting radionuclides are isolated by a BaSO4 and Fe(OH)3 co-precipitation scheme. First the gross alpha-particle activity of the sample is measured with a low-background gas-flow proportional counter, followed by radium isotopes assay by gamma-ray spectroscopy, using the same prepared sample. Gamma-ray determination of 133Ba tracer is used to assess the radium chemical recovery. The 224Ra, 226Ra, and 228Ra activities in the sample are measured through their gamma-ray-emitting decay products, 212Pb, 214Pb/214Bi, and 228Ac, respectively. In cases where 224Ra determination is required, the gamma-ray counting should be performed within 2-4 d from sample collection. To measure 226Ra activity in the sample, the gamma-ray spectroscopy can be repeated 21 d after sample preparation to ensure that 226Ra and its progeny have reached the equilibrium state. At this point, the 228Ac equilibration with parent 228Ra is already established. Analysis of aliquots of de-ionized water spiked with NIST-traceable 230Th, 224Ra, 226Ra, and 228Ra standards demonstrated the accuracy and precision of this method. Various performance evaluation samples were also assayed for gross alpha as well as radium isotope activity determination using this procedure and the results were in close agreement with the assigned values. In addition, method comparison results of actual sample analyses agreed well with the ones performed using EPA-approved procedures. With a 3-L sample aliquot and 1,000-min counting time, the average gross alpha minimum detectable concentration (MDC) was about 0.002 Bq L(-1). The average MDC's for 224Ra, 226Ra, and 228Ra were 0.034 Bq L(-1), 0.017 Bq L(-1), and 0.036 Bq L(-1), respectively, based on a 3-L sample aliquot, 85% chemical yield, 40% intrinsic Ge detector, and 1,000-min count time. This method combines and simplifies the analytical procedures and reduces labor while achieving the precision, accuracy, and minimum detection limit requirements of EPA regulations.
Parsa
Bahman
B
New Jersey Department of Health and Senior Services, Radioanalytical Services, P.O. Box 361, Trenton, NJ 08625-0361, USA. bahman.parsa@doh.state.nj.us
Obed
Reynaldo N
RN
Nemeth
William K
WK
Suozzo
Gail P
GP
D000512:Alpha Particles; D011883:Radium / Q000032:analysis*; D014875:Water Pollutants, Radioactive / Q000032:analysis*; D014881:Water Supply / Q000032:analysis*
D014875:Water Pollutants, Radioactive; D011883:Radium