Precise quantification of pulsatility is a necessity for direct comparisons of six different pediatric heart-lung machines in a neonatal CPB model.
Generation of pulsatile flow depends on an energy gradient. Surplus hemodynamic energy (SHE) is the extra hemodynamic energy generated by a pulsatile device when the adequate pulsatility is achieved. The objective of this study was to precisely quantify and compare pressure-flow waveforms in terms of surplus hemodynamic energy levels of six different pediatric heart-lung machines in a neonatal piglet model during cardiopulmonary bypass (CPB) procedures with deep hypothermic circulatory arrest (DHCA). Thirty-nine piglets (average weight, 3 kg) were subjected to CPB with a hydraulically driven physiologic pulsatile pump (PPP; n=7), Jostra-HL 20 pulsatile roller pump (Jostra-PR; n=6), Stockert Sill pulsatile roller pump (SIII-PR; n=6), Stockert Sill mast-mounted pulsatile roller pump with a miniature roller head (Mast-PR; n=7), Stockert Sill mast-mounted nonpulsatile roller pump (Mast-NP; n=7), or Stockert CAPS nonpulsatile roller pump (CAPS-NP, n=7). Once CPB was begun, each animal underwent 20 minutes of hypothermia, 60 minutes of DHCA, 10 minutes of cold reperfusion, and 40 minutes of rewarming. The pump flow rate was maintained at 150 ml x kg(-1) x min(-1) and the mean arterial pressure (MAP) at 45 mm Hg. In the pulsatile experiments, the pump rate was kept at 150 bpm and the stroke volume at 1 ml/kg. The SHE (ergs/cm3) = 1,332 ([(integral fpdt) / (integral fdt)] - MAP) was calculated at each experimental stage. During normothermic CPB (15 minutes on pump), the physiologic pulsatile pump generated the highest surplus hemodynamic energy (8563 +/- 1918 ergs/cm3, p < 0.001) compared with all other pumps. The Jostra HL-20 and Stockert Sill pulsatile roller pumps also produced adequate surplus hemodynamic energy. Nonpulsatile roller pumps and the Stockert Sill mast-mounted pulsatile roller pump did not generate any extra hemodynamic energy. During hypothermic CPB and after DHCA and rewarming, the results were extremely similar to those seen during normothermic CPB. The surplus hemodynamic energy formula is a novel method to precisely quantify different levels of pulsatility and nonpulsatility for direct and meaningful comparisons. The PPP produced the greatest surplus hemodynamic energy. Most of the pediatric pulsatile pumps (except Mast-PR) generated significant surplus hemodynamic energy. None of the nonpulsatile roller pumps generated adequate surplus hemodynamic energy.
Undar
Akif
A
Department of Pediatrics, Penn State Children's Hospital, Penn State College of Medicine, Penn State Milton S. Hershey Medical Center, Hershey, PA 17033-0850, USA.
Eichstaedt
Harald C
HC
Masai
Takafumi
T
Bigley
Joyce E
JE
Kunselman
Allen R
AR
D000818:Animals; D000831:Animals, Newborn; D001783:Blood Flow Velocity; D002315:Cardiopulmonary Bypass / Q000295:instrumentation*; D004734:Energy Metabolism; D006353:Heart-Assist Devices; D007036:Hypothermia, Induced; D011673:Pulsatile Flow; D017694:Rewarming; D013552:Swine