Beam dumps design and local radiation protection at TERA synchrotron.
The realisation of the National Center of Hadrontherapy was funded by the Italian Government in 2002. The Centre will be built in the area of Pavia (Italy). The synchrotron designed in the framework of this programme will accelerate protons and carbon ions up to 250 MeV and 400 MeV u(-1), respectively. Some of the main aspects which were taken into account in the design of the acceleration system are the patient's safety and the beam control. From this point of view an important role is played by the beam dumps in the synchrotron ring and upstream of the extraction system. In particular, an horizontal and a vertical beam dump will be installed in the synchrotron ring: the former will be used for lowering the beam intensity and the latter for beam abortion. The dump at the extraction will absorb the particles during the mounting and the falling ramps of the synchrotron magnetic cycle, thus extracting only the flat top of the ion spill. Beam dumps can produce intense fields of secondary radiation (neutrons, charged light-hadrons and photons) and high rates of induced activity, since they can absorb the beam completely. Usually they have to be shielded to protect the electronics during machine operation and to attenuate the radiation dose below the limits imposed by the law when the personnel access to the synchrotron hall. The part of the shielding design of the beam dumps concerning with the acceleration of protons was made using Monte Carlo simulations with the FLUKA code. Both induced activity and secondary radiation were taken into account. The shields against secondary radiation produced by carbon ions were designed, referring only to secondary neutrons, taking double-differential distributions from the literature as sources for the FLUKA simulations. The induced activity from carbon ions interactions was estimated analytically, using the data generated by the EPAX 2 code. The dose-equivalent rates from the induced radionuclides were calculated at 1 m from the shielded dumps, taking into account the contribution of activated components of the synchrotron ring.
Porta
A
A
CeSNEF-Centro Studi Nucleari Enrico Fermi, Polytechnic of Milan, Via Ponzio 34/3, I-20133 Milan, Italy. alessandro.porta@polimi.it
Campi
F
F
Agosteo
S
S
D003198:Computer Simulation; D017076:Computer-Aided Design; D004867:Equipment Design / Q000379:methods; D019544:Equipment Failure Analysis / Q000379:methods; D005160:Facility Design and Construction / Q000295:instrumentation / Q000379:methods*; D007558:Italy; D015233:Models, Statistical; D009010:Monte Carlo Method; D010315:Particle Accelerators / Q000295:instrumentation*; D011522:Protons; D011829:Radiation Dosage; D011834:Radiation Monitoring / Q000379:methods*; D011835:Radiation Protection / Q000295:instrumentation* / Q000379:methods*; D018570:Risk Assessment / Q000379:methods; D012307:Risk Factors; D017356:Synchrotrons / Q000295:instrumentation*
D011522:Protons