Development of a safe guidewire.
As the result of a locking phenomenon that may occur in a guidewire inside a metal puncture needle when using the Seldinger technique to insert a central venous catheter, the guidewire can break and cause an embolism. To counter this possibility we devised a guidewire with a structure that made it difficult for locking to occur and compared it to conventional guidewires. Conventional guidewires are wound lengthways with a spring. The improved version has a special multi-ply structure. A series of 100 cases were divided into two groups: group A, the conventional guidewire group; and group B, the improved guidewire group. We punctured the internal jugular vein and attempted insertion of the guidewire through the side hole of a 22-gauge metal needle. We then compared the frequency of locking and the frequency of bending of the guidewire tips that have been withdrawn. In group A, locking occurred in 72% of the cases where the guidewire was unable to be inserted, but this figure was 0% in group B. The improved guidewire has the advantage of reducing the risk of locking and of guidewire breakage.
Suzuki
Toshiyasu
T
Department of Anesthesiology, School of Medicine, Tokai University, Boseidai, Isehara 259-1193, Japan.
Ito
Kenji
K
Nishiyama
Junichi
J
Hasegawa
Keiichiro
K
Kanazawa
Masahiro
M
Fukuyama
Haruo
H
D002405:Catheterization, Central Venous / Q000295:instrumentation*; D004867:Equipment Design; D006801:Humans