title: An impulse-driven liquid-droplet deposition interface for combining LC with MALDI MS and MS/MS.
abstract: A simple and robust impulse-driven droplet deposition system was developed for off-line liquid chromatography matrix-assisted laser desorption ionization mass spectrometry (LC-MALDI MS). The system uses a solenoid operated with a pulsed voltage power supply to generate impulses that dislodge the hanging droplets from the LC outlet directly to a MALDI plate via a momentum transfer process. There is no contact between the LC outlet and the collection surface. The system is compatible with solvents of varying polarity and viscosity, and accommodates the use of hydrophobic and hydrophilic MALDI matrices. MALDI spots are produced on-line with the separation, and do not require further processing before MS analysis. It is shown that high quality MALDI spectra from 5 fmol of pyro-Glu-fibrinopeptide deposition after LC separation could be obtained using the device, indicating that there was no sample loss in the interface. To demonstrate the analytical performance of the system as a proteome analysis tool, a range of BSA digest concentrations covering about 3 orders of magnitude, from 5 fmol to 1 pmol, were analyzed by LC-MALDI quadrupole time-of-flight MS, yielding 6 and 57% amino acid sequence coverage, respectively. In addition, a complex protein mixture of an E. coli cell extract was tryptically digested and analyzed by LC-MALDI MS, resulting in the detection of a total of 409 unique peptides from 100 fractions of 15-s intervals.
authors.affiliation: Department of Chemistry, University of Alberta, Edmonton, Alberta, Canada.
authors.forename: J Bryce
authors.initials: JB
authors.lastname: Young
authors.forename: Liang
authors.initials: L
authors.lastname: Li
mesh_terms: D002853:Chromatography, Liquid / Q000295:instrumentation* / Q000379:methods; D004867:Equipment Design; D019544:Equipment Failure Analysis; D017022:Flow Injection Analysis / Q000295:instrumentation / Q000379:methods; D044085:Microfluidics / Q000295:instrumentation* / Q000379:methods; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity; D012996:Solutions; D013048:Specimen Handling / Q000295:instrumentation* / Q000379:methods; D019032:Spectrometry, Mass, Matrix-Assisted Laser Desorption-Ionization / Q000295:instrumentation* / Q000379:methods
chemical_list: D012996:Solutions