A new device producing manual sternal compression with thoracic constraint for cardiopulmonary resuscitation.
OBJECTIVE
Blood flow during conventional cardiopulmonary resuscitation (CPR) is usually less than adequate to sustain vital organ perfusion. A new chest compression device (LifeBelt) which compresses both the sternum and the lateral thoraces (compression and thoracic constraint) has been developed. The device is light weight, portable, manually powered and mechanically advantaged to minimize user fatigue. The purpose of this study was to evaluate the mechanism of blood flow with the device, determine the optimal compression force and compare the device to standard manual CPR in a swine arrest model.


METHODS
Following anesthesia and instrumentation, intravascular contrast injections were performed in four animals and the performance characteristics of the device were evaluated in eight animals. In a comparative outcome study, 42 anesthetized and instrumented swine were randomized to receive LifeBelt or manual CPR. Ventricular fibrillation (VF) was induced electrically and was untreated for 7.5 min. After 7.5 min, countershocks were administered and chest compressions initiated. Pulseless electrical activity (PEA) was observed after one to three shocks in all animals. CPR was continued until restoration of spontaneous circulation (ROSC) or for 10 min after the first shock. If ROSC had not occurred within 5 min of beginning CPR, 0.01 mg/kg of epinephrine (adrenaline) was administered. During CPR, peak systolic aortic pressure (Ao), diastolic coronary perfusion pressure (CPP-diastolic aortic minus diastolic right pressure) and end-tidal CO(2) were measured.


RESULTS
Angiographic studies demonstrated cardiac compression as the mechanism of blood flow. Optimal performance, determined by coronary perfusion pressure, was observed at a sternal force of 100-130 lb (45-59 kg). In the comparative trial, significant differences in the measured CPP were observed between LifeBelt and manual CPR both at 1 min (15+/-8 mmHg versus 10+/-6 mmHg, p<0.05) and 5 min (17+/-4 mmHg versus 13+/-7 mmHg, p<0.02) of chest compression. A greater (p<0.05) ETCO(2), a marker of cardiac output and systemic perfusion, was observed with LifeBelt CPR (20+/-7 mmHg) than with manual CPR (15+/-5 mmHg) at 1 min. Peak Ao pressures were not different between methods. With the device, 86% of animals were resuscitated compared to 76% in the manual group.


CONCLUSIONS
Blood flow with the LifeBelt device is primarily the result of cardiac compression. At a sternal force of 100-130 lb (45-59 kg), the device produces greater CPP than well-performed manual CPR during resuscitation from prolonged VF.
Niemann
James T
JT
Department of Emergency Medicine, Division of Cardiology, Harbor-UCLA Medical Center, Box 21, 1000 West Carson Street, Torrance, CA 90509, USA. jniemann@emedharbor.edu
Rosborough
John P
JP
Kassabian
Leo
L
Salami
Bobak
B
D000818:Animals; D001022:Aortic Valve Insufficiency / Q000503:physiopathology; D016887:Cardiopulmonary Resuscitation / Q000295:instrumentation*; D003326:Coronary Circulation; D005260:Female; D006323:Heart Arrest / Q000628:therapy*; D008297:Male; D011312:Pressure / Q000009:adverse effects; D013249:Sternum; D013552:Swine; D014693:Ventricular Fibrillation / Q000628:therapy*