Cherry-picking in an orchard: unattended LC/MS analysis from an autosampler with >32,000 samples online.
The 1,536-well microplate format has widely supplanted the 384-well microplate format for high-throughput screening and for IC(50) assays. Previously, liquid chromatography/mass spectrometry (LC/MS) analyses of such samples required manual transfers of the wells of interest from a 1,536-well plate into a 384-well plate. Because this manual transfer introduced a source of potential error, it became clear that a more appropriate solution would be to sample directly from the 1,536-well plates. Currently, commercially available 1,536-well plate auto samplers are not compatible with Waters LC/MS systems. The authors have modified their CTC PAL autosampler to support injection from up to twenty-four 1,536-well plates. This allows them to cherry-pick any sample from up to 36,864 wells on the autosampler. Because of its success at this Institute, sampling from 1,536-well plates has not only become the preferred method for LC/MS analysis from IC(50) plates but also become the standard format used for the handling of and the sampling from large combinatorial libraries.
Ari
Nader
N
The Genomics Institute of the Novartis Research Foundation, San Diego, California 92121, USA. nari@gnf.org
Westling
Lucas
L
Isbell
John
J
D002853:Chromatography, Liquid / Q000295:instrumentation* / Q000379:methods; D015203:Reproducibility of Results; D013056:Spectrophotometry, Ultraviolet / Q000295:instrumentation* / Q000379:methods