Friction produced by types of elastomeric ligatures in treatment mechanics with the preadjusted appliance.
The objective was to compare the frictional forces generated by new nonconventional passive elastomeric ligatures (NCL) and conventional elastomeric ligatures (CL) under dry conditions. An experimental model reproducing the right buccal segment of the upper arch and consisting of five stainless steel 0.022-inch preadjusted brackets (from the second premolar through the central incisor) was used to assess both static and kinetic frictional forces produced by NCL and CL. The frictional forces generated by the 0.019 x 0.025-inch stainless steel wire with the two types of elastomeric ligatures were recorded by sliding the wire into the aligned brackets. The friction produced by the 0.014-inch superelastic nickel titanium wire was evaluated both in the presence of aligned brackets and of three-mm misaligned canine bracket. The amount of both static and kinetic frictions were minimal (<10 g) in the NCL group in the presence of aligned brackets with both types of wires, whereas it ranged from a minimum of 95.6 g for the 0.014-inch superelastic nickel titanium wire to a maximum of 590.7 g for the 0.019 x 0.025-inch stainless steel wire when using CL. The amount of both static and kinetic frictions in the presence of a misaligned canine bracket in the NCL group were less than half of that shown by the CL group. A recently developed passive ligature system is able to produce significantly lower levels of frictional forces in vitro when compared with conventional elastomeric modules.
Baccetti
Tiziano
T
Department of Orthodontics, The University of Florence, Florence, Italy. t.baccetti@odonto.unifi.it
Franchi
Lorenzo
L
D020724:Elastomers / Q000737:chemistry*; D017276:Friction; D007700:Kinetics; D008026:Ligation; D009967:Orthodontic Appliances; D016910:Orthodontic Brackets; D009969:Orthodontic Wires; D018709:Statistics, Nonparametric
D020724:Elastomers