Operational specifications of the laser illuminated track etch scattering dosemeter reader.
The personnel dosimetry operations team at the Los Alamos National Laboratory (LANL) has accepted the laser illuminated track etch scattering (LITES) dosemeter reader into its suite of radiation dose measurement instruments. The LITES instrument transmits coherent light from a He-Ne laser through the pertinent track etch foil and a photodiode measures the amount of light scattered by the etched tracks. A small beam stop blocks the main laser light, while a lens refocuses the scattered light into the photodiode. Three stepper motors in the current LITES system are used to position a carousel that holds 36 track etch dosemeters (TEDs). Preliminary work with the LITES system demonstrated the device had a linear response in counting foils subjected to exposures up to 50 mSv (5.0 rem). The United States Department of Energy requires that the annual general employee dose not exceed 50 mSv (5.0 rem). On a regular basis, LANL uses the Autoscan-60 reader system (Thermo Electron Corp.) for counting track etch dosemeters. However, LANL uses a 15 h etch process for CR-39 dosemeters, and this produces more and larger track etch pits than the 6 h etch used by many institutions. Therefore, LANL only uses the Autoscan-60 for measuring neutron dose equivalent up to exposure levels of approximately 3 mSv (300 mrem). The LITES system has a measured lower limit of detection of approximately 0.6 mSv (60 mrem), and it has a correlation coefficient of R (2) = 0.99 over an exposure range up to 500 mSv (50.0 rem). A series of blind studies were done using three methods: the Autoscan-60 system, manual counting by optical microscope and the LITES instrument. A collection of track etch dosemeters of unknown neutron dose equivalent (NDE) were analysed using the three methods, and the performance coefficient (PC) was calculated when the NDE became known. The Autoscan-60 and optical microscope methods had a combined PC = 0.171, and the LITES instrument had a PC = 0.194, where a PC less than or equal to 0.300 is considered satisfactory.
Moore
M E
ME
Los Alamos National Laboratory, Los Alamos, NM 87545, USA. memoore@lanl.gov
Gepford
H J
HJ
Hoffman
J M
JM
McKeever
R J
RJ
Devine
R T
RT
D004307:Dose-Response Relationship, Radiation; D004867:Equipment Design; D019544:Equipment Failure Analysis; D007834:Lasers; D011829:Radiation Dosage; D011835:Radiation Protection / Q000295:instrumentation*; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity; D013499:Surface Properties; D013819:Thermoluminescent Dosimetry / Q000295:instrumentation* / Q000379:methods