The 12B counter: an active dosemeter for high-energy neutrons.
High-energy accelerators can produce strong time-structured radiation fields. Such dose shots are generated at linear machines with low duty cycles as well as at circular machines when complete fills are instantaneously lost. The main dose component behind thick shielding is due to high-energy neutrons occurring at that time structure. Dosemeters based on Geiger-Mueller tubes or proportional counters fail here completely. The 12B counter, a novel dosemeter made of a plastic scintillator using carbon activation for event-like exposure, has been introduced. High-energy neutrons activate the carbon nuclei by three inelastic reactions. The decay patterns with half-lives between 20 ms and 20 min can be exploited depending on the time structure of the radiation field. The response of the 12B counter was measured along with some other dosemeters, both active and passive, in the radiation field behind the lateral concrete shielding of a 7.5 GeV proton transfer line.
Leuschner
A
A
Deutsches Elektronen-Synchrotron DESY, Notkestrasse 85, D-22607 Hamburg, Germany. albrecht.leuschner@desy.de
D001895:Boron / Q000032:analysis*; D004867:Equipment Design; D019544:Equipment Failure Analysis; D005214:Fast Neutrons; D010315:Particle Accelerators / Q000295:instrumentation*; D011829:Radiation Dosage; D011835:Radiation Protection / Q000295:instrumentation* / Q000379:methods; D011868:Radioisotopes / Q000032:analysis*; D011874:Radiometry / Q000295:instrumentation* / Q000379:methods; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity
D011868:Radioisotopes; D001895:Boron