The crystallography beamline I711 at MAX II.
A new X-ray crystallographic beamline is operational at the MAX II synchrotron in Lund. The beamline has been in regular use since August 1998 and is used both for macro- and small molecule diffraction as well as powder diffraction experiments. The radiation source is a 1.8 T multipole wiggler. The beam is focused vertically by a bendable mirror and horizontally by an asymmetrically cut Si(111) monochromator. The wavelength range is 0.8-1.55 A with a measured flux at 1 A of more than 10(11) photons s(-1) in 0.3 mm x 0.3 mm at the sample position. The station is currently equipped with a Mar345 imaging plate, a Bruker Smart 1000 area CCD detector and a Huber imaging-plate Guinier camera. An ADSC 210 area CCD detector is planned to be installed during 2000.
Cerenius
Y
Y
Department of Molecular Biophysics, Center for Chemistry and Chemical Engineering, Lund University, PO Box 124, S221 00 Lund, Sweden. yngve.cerenius@mbfys.lu.se
Ståhl
K
K
Svensson
L A
LA
Ursby
T
T
Oskarsson
A
A
Albertsson
J
J
Liljas
A
A