Evaluation of the COBAS AmpliPrep-total nucleic acid isolation-COBAS TaqMan hepatitis B virus (HBV) quantitative test and comparison to the VERSANT HBV DNA 3.0 assay.
Quantitative detection of hepatitis B virus (HBV) in serum or plasma has become the most direct and reliable method for monitoring chronic hepatitis B. Here, we report the performance characteristics of a real-time PCR hepatitis B DNA quantitative assay, the COBAS TaqMan (CTM) HBV test (Roche Diagnostics, Meylan, France), in combination with an automated DNA extraction on the COBAS AmpliPrep (CAP) instrument using the total nucleic acid isolation kit (TNAI kit), a generic reagent for nucleic acid isolation (both from Roche Diagnostics). The linearity, accuracy, and specificity of the CAP-TNAI-CTM HBV test were evaluated using various reference panels and standards (HBV panel 2004 from Quality Control for Molecular Diagnostics, OptiQuant HBV panel from AcroMetrix, WHO International Standard for HBV, and Teragenix hepatitis B genotype panel). Quantitative results show that the CAP-TNAI-CTM HBV test performed well with respect to linearity, accuracy, and reproducibility from at least 100 to 500,000 HBV DNA IU/ml. Based on the log(10) IU of HBV DNA/ml measured, the intra-assay variation ranged from 2.49% to 8.46% and the interassay variation ranged from 1.88% to 7.83%. The test was extremely sensitive and could detect samples containing HBV DNA below the reported quantification threshold (<30 IU/ml). All HBV genotypes were correctly amplified, and no cross-contamination occurred during the automated sample preparation. In addition, 402 human serum samples were tested comparatively to the VERSANT HBV DNA 3.0 assay (bDNA; Bayer Diagnostics, Puteaux, France). The viral load results of the CAP-TNAI-CTM test and bDNA were significantly correlated, but the agreement between the two tests was poor, with large differences between results for individual samples. The hands-on time was estimated to be reduced from 2.30 h with bDNA to 45 min with the CAP-TNAI-CTM test, and up to 84 samples were completely processed within a working day. Overall, the performance characteristics of the CAP-TNAI-CTM test demonstrated that it provides a high-throughput sensitive and reliable method for quantitation of HBV DNA levels in the routine molecular laboratory.
Ronsin
Christophe
C
Laboratoire L.C.L., Service de Biologie Moléculaire, 78 Avenue de Verdun, BP 110, 94208 Ivry sur Seine, France. c.ronsin@lablcl.com.
Pillet
Anne
A
Bali
Corinne
C
Denoyel
Gérard-Antoine
GA
D004279:DNA, Viral / Q000097:blood*; D006509:Hepatitis B / Q000175:diagnosis* / Q000821:virology; D006515:Hepatitis B virus / Q000145:classification / Q000235:genetics / Q000302:isolation & purification*; D006801:Humans; D021141:Nucleic Acid Amplification Techniques / Q000295:instrumentation / Q000379:methods; D016133:Polymerase Chain Reaction / Q000379:methods; D011933:Reagent Kits, Diagnostic; D019914:Taq Polymerase / Q000378:metabolism
D004279:DNA, Viral; D011933:Reagent Kits, Diagnostic; D019914:Taq Polymerase