Rotating sample system: an equivalent of a rotating electrode for microliter samples.
Voltammetric experiments in sample droplets of 20 microL volume were carried out with gold ring electrodes microfabricated on Pyrex substrates. The droplet studied was centered and kept in place by a hydrophobic ring deposited on the substrate around the ring electrode, which also ensured that each sample assume a semispherical shape. A small hole in the center of the substrate filled with an agar gel membrane served as a junction. A mild jet of humidified nitrogen gas directed tangentially at the droplet caused it to rotate at a high rate. Cyclic voltammetry and constant-potential electrolysis of potassium ferricyanide were used to characterize this rotating sample system and to calculate diffusion layer thickness. The results clearly demonstrate that rotating of a semispherical microsample placed above a stationary ring electrode with a mild gas jet can be as effective as a rotating electrode system. As a practical example, voltammetric stripping analysis of 20-120 pmol mercuric nitrate samples was performed with a calibration of good linearity (r2 = 0.988) and a time constant for exhaustive electrolysis of approximately 2 min.
Cserey
A
A
Department of Biomedical Engineering, Case Western Reserve University, Cleveland, Ohio 44106, USA.
Gratzl
M
M
D004563:Electrochemistry; D017669:Mercury Compounds / Q000032:analysis; D008839:Microelectrodes; D046210:Microfluidic Analytical Techniques / Q000379:methods*; D009566:Nitrates / Q000032:analysis; D012399:Rotation
D017669:Mercury Compounds; D009566:Nitrates; C029948:mercury nitrate