title: Roche/BIOTECON Diagnostics LightCycler foodproof L. monocytogenes detection kit in combination with ShortPrep foodproof II Kit. Performance-Tested Method 070401.
abstract: A method was developed for the detection of L. monocytogenes in food based on real-time polymerase chain reaction (PCR). This advanced PCR method was designed to reduce the time needed to achieve results from PCR reactions and to enable the user to monitor the amplification of the PCR product simultaneously, in real-time. After DNA isolation using the Roche/BIOTECON Diagnostics ShortPrep foodproof II Kit (formerly called Listeria ShortPrep Kit) designed for the rapid preparation of L. monocytogenes DNA for direct use in PCR, the real-time detection of L. monocytogenes DNA is performed by using the Roche/BIOTECON Diagnostics LightCycler foodproof L. monocytogenes Detection Kit. This kit provides primers and hybridization probes for sequence-specific detection, convenient premixed reagents, and different controls for reliable interpretation of results. For repeatability studies, 20 different foods, covering the 15 food groups recommended from the AOAC Research Institute (AOAC RI) for L. monocytogenes detection were analyzed: raw meats, fresh produce/vegetables, processed meats, seafood, egg and egg products, dairy (cultured/noncultured), spices, dry foods, fruit/juices, uncooked pasta, nuts, confectionery, pet food, food dyes and colorings, and miscellaneous. From each food 20, samples were inoculated with a low level (1-10 colony-forming units (CFU)/25 g) and 20 samples with a high level (10-50 CFU/25 g) of L. monocytogenes. Additionally, 5 uninoculated samples were prepared from each food. The food samples were examined with the test kits and in correlation with the cultural methods according to U.S. Food and Drug Administration (FDA) Bacteriological Analytical Manual (BAM) or U.S. Department of Agriculture (USDA)/Food Safety and Inspection Service (FSIS) Microbiology Laboratory Guidebook. After 48 h of incubation, the PCR method in all cases showed equal or better results than the reference cultural FDA/BAM or USDA/FSIS methods. Fifteen out of 20 tested food types gave exactly the same amount of positive samples for both methods in both inoculation levels. For 5 out of 20 foodstuffs, the PCR method resulted in more positives than the reference method after 48 h of incubation. Following AOAC RI definition, these were false positives because they were not confirmed by the reference method (false-positive rate for low inoculated foodstuffs: 5.4%; for high inoculated foodstuffs: 7.1%). Without calculating these unconfirmed positives, the PCR method showed equal sensitivity results compared to the alternative method. With the unconfirmed PCR-positives included into the calculations, the alternative PCR method showed a higher sensitivity than the microbiological methods (low inoculation level: 100 vs 98.0%; sensitivity rate: 1; high inoculation level: 99.7 vs 97.7%; sensitivity rate, 1). All in-house and independently tested uninoculated food samples were negative for L. monocytogenes. The ruggedness testing of both ShortPrep foodproof II Kit and Roche/BIOTECON LightCycler foodproof L. monocytogenes Detection Kit showed no noteworthy influences to any variation of the parameters component concentration, apparatus comparison, tester comparison, and sample volumes. In total, 102 L. monocytogenes isolates (cultures and pure DNA) were tested and detected for the inclusivity study, including all isolates claimed by the AOAC RI. The exclusivity study included 60 non-L. monocytogenes bacteria. None of the tested isolates gave a false-positive result; specificity was 100%. Three different lots were tested in the lot-to-lot study. All 3 lots gave equal results. The stability study was subdivided into 3 parts: long-term study, stress test, and freeze-defrost test. Three lots were tested in 4 time intervals within a period of 13 months. They all gave comparable results for all test intervals. For the stress test, LightCycler L. monocytogenes detection mixes were stored at different temperatures and tested at different time points during 1 month. Stable results were produced at all storage temperatures. The freeze-defrost analysis showed no noteworthy aggravation of test results. The independent validation study examined by Campden and Chorleywood Food Research Association Group (CCFRA) demonstrated again that the LightCycler L. monocytogenes detection system shows a comparable sensitivity to reference methods. With both the LightCycler PCR and BAM methods, 19 out of 20 inoculated food samples were detected. The 24 h PCR results generated by the LightCycler system corresponded directly with the FDA/BAM culture results. However, the 48 h PCR results did not relate exactly to the FDA/BAM results, as one sample found to be positive by the 48 h PCR could not be culturally confirmed and another sample which was negative by the 48 h PCR was culturally positive.
authors.affiliation: BIOTECON Diagnostics GmbH, Hermannswerder Haus 17, 14473 Potsdam, Germany. bjunge@bc-diagnostics.com
authors.forename: Benjamin
authors.initials: B
authors.lastname: Junge
authors.forename: Kornelia
authors.initials: K
authors.lastname: Berghof-Jäger
mesh_terms: D004269:DNA, Bacterial / Q000235:genetics; D005504:Food Analysis / Q000295:instrumentation*; D005506:Food Contamination / Q000032:analysis*; D005516:Food Microbiology; D008089:Listeria monocytogenes / Q000737:chemistry*; D011933:Reagent Kits, Diagnostic; D012015:Reference Standards; D015203:Reproducibility of Results; D020133:Reverse Transcriptase Polymerase Chain Reaction; D014481:United States; D014486:United States Food and Drug Administration
chemical_list: D004269:DNA, Bacterial; D011933:Reagent Kits, Diagnostic