Evaluation of quality of commercial pedometers.
BACKGROUND
The purpose of this study was to: 1) evaluate the quality of promotional pedometers widely distributed through cereal boxes at the time of the 2004 Canada on the Move campaign; and 2) establish a battery of testing protocols to provide direction for future consensus on industry standards for pedometer quality.


METHODS
Fifteen Kellogg's* Special K* Step Counters (K pedometers or K; manufactured for Kellogg Canada by Sasco, Inc.) and 9 Yamax pedometers (Yamax; Yamax Corporation, Tokyo, Japan) were tested with 9 participants accordingly: 1) 20 Step Test; 2) treadmill at 80m x min(-1) (3 miles x hr(-1)) and motor vehicle controlled conditions; and 3) 24-hour free-living conditions against an accelerometer criterion.


RESULTS
Fifty-three percent of the K pedometers passed the 20 Step Test compared to 100% of the Yamax. Mean absolute percent error for the K during treadmill walking was 24.2+/-33.9 vs. 3.9+/-6.6% for the Yamax. The K detected 5.7-fold more non-steps compared to the Yamax during the motor vehicle condition. In the free-living condition, mean absolute percent error relative to the ActiGraph was 44.9+/-34.5% for the K vs. 19.5+/-21.2% for the Yamax.


CONCLUSIONS
K pedometers are unacceptably inaccurate. We suggest that research grade pedometers: 1) be manufactured to a sensitivity threshold of 0.35 Gs; 2) detect +/-1 step error on the 20 Step Test (i.e., within 5%); 3) detect +/-1% error most of the time during treadmill walking at 80m x min(-1) (3 miles x hr(-1)); as well as, 4) detect steps/day within 10% of the ActiGraph at least 60% of the time, or be within 10% of the Yamax under free-living conditions.
Tudor-Locke
Catrine
C
Walking Research Laboratory, Department of Exercise and Wellness, Arizona State University, Mesa, AZ 85212-0180, USA.
Sisson
Susan B
SB
Lee
Sarah M
SM
Craig
Cora L
CL
Plotnikoff
Ronald C
RC
Bauman
Adrian
A