A flow cytometric method for detection and enumeration of low-level, residual red blood cells in platelets and mononuclear cell products.
BACKGROUND
Conventional automated cell counters cannot accurately count residual red blood cells (rRBCs) that are often present in various blood products. A two-color flow cytometric method (FC) was validated for detecting and enumerating rRBCs in platelets (PLTs) and mononuclear cell (MNC) products.


METHODS
PLT and MNC products for PLTs (CD61-fluorescein isothiocyanate) and rRBC (anti-glycophorin A-phycoerythrin) were double stained, and data were acquired with a flow cytometer. Assay linearity, accuracy, and precision were assessed with a standard-dilution series of rRBCs. This assay was used to determine the rRBCs of apheresis PLTs collected with Trima Accel (Gambro BCT) and MNC products collected with the COBE Spectra (Gambro BCT).


RESULTS
The linear range of this assay in PLT and MNC products is 10 to 2000 RBCs per microL (R2=0.994). FC had a mean intraassay coefficient of variation of 11.8 percent at 34 RBCs per microL. A standard clinical hematology analyzer overestimated rRBCs in MNC products by 1.59x10(5)+/-0.7x10(5) RBCs per microL. Apheresis PLTs had a median of 17.4 RBCs per microL, with 99.0 percent containing fewer than 90.0 RBCs per microL.


CONCLUSIONS
This method for determining rRBCs in blood products is accurate and repeatable with a lower limit of detection adequate to assess currently available blood products. FC should be considered for determining rRBCs in MNC products.
Santana
Julie M
JM
R&D, Gambro BCT, Inc., Lakewood, Colorado 80215, USA.
Dumont
Larry J
LJ
D004906:Erythrocyte Count; D005434:Flow Cytometry / Q000379:methods*; D006801:Humans; D007937:Leukapheresis / Q000592:standards*; D007958:Leukocyte Count; D007963:Leukocytes, Mononuclear; D010976:Platelet Count; D010983:Plateletpheresis / Q000592:standards*