Technical aspects of the Naval Dosimetry Center quality assurance programme.
The purpose of this paper is to describe the technical aspects of the Naval Dosimetry Center (NDC) quality programme. The Navy has been formally monitoring personnel for occupational exposure to ionising radiation since at least 1946. The current system, the DT-702/PD, is the Harshaw 8840 holder and 8841 card. New card and holder checks are performed to verify that the correct LiF elements and holder filters are in the correct location and are of the correct composition. Element correction coefficient (ECC) magnitude and repeatability are also verified. Several quality assurance parameters are checked by a specially designed shipping machine. Calibration cards are used to calibrate each reader and quality control cards are inserted throughout a group of field cards to verify reader operation during the read process. The success of the programme is measured by annual proficiency tests administered by the National Voluntary Laboratory Accreditation Programme and Pacific Northwest National Laboratories.
St John
T J
TJ
The US Naval Dosimetry Center, Bethesda, MD, USA. TJStjohn@mar.med.navy.mil
Cassata
J R
JR
Blake
P K
PK
Wallace
W H
WH
Minniti
R
R
D002138:Calibration; D008889:Military Personnel; D016273:Occupational Exposure / Q000032:analysis*; D011785:Quality Assurance, Health Care / Q000379:methods* / Q000458:organization & administration*; D011829:Radiation Dosage; D013673:Technology Assessment, Biomedical; D013819:Thermoluminescent Dosimetry / Q000295:instrumentation* / Q000379:methods* / Q000592:standards; D014481:United States