Development of a multiplexed interface for capillary electrophoresis-electrospray ion trap mass spectrometry.
A four-channel multiplexed electrospray capillary electrophoresis interface has been developed. This new interface permits up to four capillary electrophoresis columns to be sampled sequentially by means of a stepper motor and a notched rotating plate assembly, which at any instant occludes all but a single sprayer. In this design, four sheath liquid electrospray probes are oriented in a circular array situated 90 degrees relative to one another. The rotating metal disk, which contains a one-quarter notch, is mounted to the stepper motor assembly and is located between the sprayers and the entrance aperture of an ion trap mass spectrometer. By using the data acquisition signal from the ion trap mass spectrometer, the scan event is synchronized with the rotation of the metal disk. With this device, four discrete sample streams can be simultaneously analyzed, resulting in a 4-fold increase in analytical throughput.
Li
Fu-An
FA
Department of Chemistry, National Taiwan University, Taipei, Taiwan, ROC.
Wu
Ming-Chi
MC
Her
Guor-Rong
GR
D019075:Electrophoresis, Capillary / Q000295:instrumentation / Q000379:methods; D004867:Equipment Design; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity; D021241:Spectrometry, Mass, Electrospray Ionization / Q000295:instrumentation* / Q000379:methods; D013499:Surface Properties; D013997:Time Factors