Management of pediatric radiation dose using Philips fluoroscopy systems DoseWise: perfect image, perfect sense.
Although image quality (IQ) is the ultimate goal for accurate diagnosis and treatment, minimizing radiation dose is equally important. This is especially true when pediatric patients are examined, because their sensitivity to radiation-induced cancer is two to three times greater than that of adults. DoseWise is an ALARA-based philosophy within Philips Medical Systems that is active at every level of product design. It encompasses a set of techniques, programs and practices that ensures optimal IQ while protecting people in the X-ray environments. DoseWise methods include management of the X-ray beam, less radiation-on time and more dose information for the operator. Smart beam management provides automatic customization of the X-ray beam spectrum, shape, and pulse frequency. The Philips-patented grid-controlled fluoroscopy (GCF) provides grid switching of the X-ray beam in the X-ray tube instead of the traditional generator switching method. In the examination of pediatric patients, DoseWise technology has been scientifically documented to reduce radiation dose to <10% of the dose of traditional continuous fluoroscopy systems. The result is improved IQ at a significantly lower effective dose, which contributes to the safety of patients and staff.
Stueve
Dick
D
Philips Medical Systems North America, 22100 Bothell-Everett Highway, Bothell, WA 98021-8431, USA. Dick.Stueve@philips.com
D001822:Body Burden; D004781:Environmental Exposure / Q000517:prevention & control*; D004867:Equipment Design; D019544:Equipment Failure Analysis; D005471:Fluoroscopy / Q000295:instrumentation*; D010372:Pediatrics / Q000295:instrumentation*; D011832:Radiation Injuries / Q000517:prevention & control*; D011835:Radiation Protection / Q000295:instrumentation* / Q000379:methods; D015641:Radiography, Interventional / Q000295:instrumentation*; D017751:Safety Management / Q000379:methods; D014481:United States