Medical diagnostic applications and sources.
The ways in which ultrasound is used in medical diagnosis are reviewed, with particular emphasis on the ultrasound source (probe) and implications for acoustic exposure. A brief discussion of the choice of optimum frequency for various target depths is followed by a description of the general features of diagnostic ultrasound probes, including endo-probes. The different modes of diagnostic scanning are then discussed in turn: A-mode, M-mode, B-mode, three-dimensional (3D) and 4D scanning, continuous wave (CW) Doppler, pulse-wave spectral Doppler and Doppler imaging. Under the general heading of B-mode imaging, there are individual descriptions of the principles of chirps and binary codes, B-flow, tissue harmonic imaging and ultrasound contrast agent-specific techniques. Techniques for improving image quality within the constraints of real-time operation are discussed, including write zoom, parallel beam forming, spatial compounding and multiple zone transmission focusing, along with methods for reducing slice thickness. At the end of each section there is a summarising comment on the basic features of the acoustic output and its consequences for patient safety.
Whittingham
T A
TA
Regional Medical Physics Department, Newcastle General Hospital, Newcastle upon Tyne NE4 6BE, UK. t.a.whittingham@ncl.ac.uk
D007089:Image Enhancement / Q000295:instrumentation* / Q000379:methods*; D007090:Image Interpretation, Computer-Assisted / Q000295:instrumentation* / Q000379:methods*; D021621:Imaging, Three-Dimensional / Q000295:instrumentation / Q000379:methods; D014463:Ultrasonography / Q000295:instrumentation* / Q000379:methods* / Q000639:trends