Synthetic aperture ultrasound imaging.
The paper describes the use of synthetic aperture (SA) imaging in medical ultrasound. SA imaging is a radical break with today's commercial systems, where the image is acquired sequentially one image line at a time. This puts a strict limit on the frame rate and the possibility of acquiring a sufficient amount of data for high precision flow estimation. These constrictions can be lifted by employing SA imaging. Here data is acquired simultaneously from all directions over a number of emissions, and the full image can be reconstructed from this data. The paper demonstrates the many benefits of SA imaging. Due to the complete data set, it is possible to have both dynamic transmit and receive focusing to improve contrast and resolution. It is also possible to improve penetration depth by employing codes during ultrasound transmission. Data sets for vector flow imaging can be acquired using short imaging sequences, whereby both the correct velocity magnitude and angle can be estimated. A number of examples of both phantom and in vivo SA images will be presented measured by the experimental ultrasound scanner RASMUS to demonstrate the many benefits of SA imaging.
Jensen
Jørgen Arendt
JA
Center for Fast Ultrasound Imaging, Ørsted DTU, Building 348, Technical University of Denmark, DK-2800 Lyngby, Denmark. jaj@oersted.dtu.dk <jaj@oersted.dtu.dk>
Nikolov
Svetoslav Ivanov
SI
Gammelmark
Kim Løkke
KL
Pedersen
Morten Høgholm
MH
D001158:Arteries / Q000000981:diagnostic imaging* / Q000502:physiology*; D001783:Blood Flow Velocity / Q000502:physiology*; D004867:Equipment Design; D019544:Equipment Failure Analysis; D006801:Humans; D007089:Image Enhancement / Q000379:methods*; D007090:Image Interpretation, Computer-Assisted / Q000379:methods*; D014463:Ultrasonography / Q000295:instrumentation* / Q000379:methods* / Q000639:trends