High frame rate imaging system for limited diffraction array beam imaging with square-wave aperture weightings.
A general-purpose high frame rate (HFR) medical imaging system has been developed. This system has 128 independent linear transmitters, each of which is capable of producing an arbitrary broadband (about 0.05-10 MHz) waveform of up to +/- 144 V peak voltage on a 75-ohm resistive load using a 12-bit/40-MHz digital-to-analog converter. The system also has 128 independent, broadband (about 0.25-10 MHz), and time-variable-gain receiver channels, each of which has a 12-bit/40-MHz analog-to-digital converter and up to 512 MB of memory. The system is controlled by a personal computer (PC), and radio frequency echo data of each channel are transferred to the same PC via a standard USB 2.0 port for image reconstructions. Using the HFR imaging system, we have developed a new limited-diffraction array beam imaging method with square-wave aperture voltage weightings. With this method, in principle, only one or two transmitters are required to excite a fully populated two-dimensional (2-D) array transducer to achieve an equivalent dynamic focusing in both transmission and reception to reconstruct a high-quality three-dimensional image without the need of the time delays of traditional beam focusing and steering, potentially simplifying the transmitter subsystem of an imager. To validate the method, for simplicity, 2-D imaging experiments were performed using the system. In the in vitro experiment, a custom-made, 128-element, 0.32-mm pitch, 3.5-MHz center frequency linear array transducer with about 50% fractional bandwidth was used to reconstruct images of an ATS 539 tissue-mimicking phantom at an axial distance of 130 mm with a field of view of more than 90 degrees. In the in vivo experiment of a human heart, images with a field of view of more than 90 degrees at 120-mm axial distance were obtained with a 128-element, 2.5-MHz center frequency, 0.15-mm pitch Acuson V2 phased array. To ensure that the system was operated under the limits set by the U.S. Food and Drug Administration, the mechanical index, thermal index, and acoustic output were measured. Results show that higher-quality images can be reconstructed with the square-wave aperture weighting method due to an increased penetration depth as compared to the exact weighting method developed previously, and a frame rate of 486 per second was achieved at a pulse repetition frequency of about 5348 Hz for the human heart.
Lu
Jian-Yu
JY
Ultrasound Laboratory, Department of Bioengineering, The University of Toledo, Toledo, OH 43606, USA. jilu@eng.utoledo.edu
Cheng
Jiqi
J
Wang
Jing
J
D003199:Computer Systems; D004867:Equipment Design; D019544:Equipment Failure Analysis; D007089:Image Enhancement / Q000295:instrumentation* / Q000379:methods; D007090:Image Interpretation, Computer-Assisted / Q000295:instrumentation* / Q000379:methods; D019047:Phantoms, Imaging; D012031:Refractometry / Q000295:instrumentation* / Q000379:methods; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity; D012815:Signal Processing, Computer-Assisted / Q000295:instrumentation*; D018511:Systems Integration; D014159:Transducers; D014463:Ultrasonography / Q000295:instrumentation*