A second look at the Palmes' diffusive sampler.
The Palmes' tube, the first diffusive sampler incorporating a fixed path length, has received wide usage for the sampling of a large number of gaseous pollutants. But despite numerous previous studies, questions remain regarding the accuracy of these inexpensive, simple-to-construct, open-ended samplers. Here the mass transfer resistance in a Palmes' diffusive sampler was measured using the loss of cyclohexane from a Palmes' tube containing liquid cyclohexane at its base. The average loss rates, at factorial combinations of five air incidence angles evenly spaced from 270 degrees to 90 degrees, and five air speeds from 0.5 m/sec to 2.5 m/sec ranged from 46% to 121% higher than rates calculated from the physical dimensions of the sampler, proving the need to calibrate these samplers rather than relying on a theoretical calculation. The mass transfer resistance was nearly constant when the airflow was perpendicular to the sampler and sufficiently high to avoid stagnation, a finding that may explain the widespread acceptance of the results obtained using this sampler.
Busbin
Dainnya D
DD
Arnold School of Public Health, University of South Carolina, Columbia, SC, USA.
Feigley
Charles E
CE
Underhill
Dwight W
DW
Salzberg
Deborah
D
D000393:Air Pollutants / Q000032:analysis; D003510:Cyclohexanes / Q000032:analysis; D004784:Environmental Monitoring / Q000295:instrumentation*; D014919:Wind
D000393:Air Pollutants; D003510:Cyclohexanes; C506365:Cyclohexane