title: Determination of fluoroquinolones in milk samples by postcolumn derivatization liquid chromatography with luminescence detection.
abstract: A liquid chromatography (LC) method with luminescence detection for the determination of eight quinolone antibiotics is reported. The system encompasses three consecutive steps: (a) chromatographic separation using reverse-phase mode (RP-LC), (b) postcolumn derivatization reaction, and (c) luminescence detection by monitoring fluorescence (FL) and time-resolved (TR) signals. The derivatization step is based on the reaction between quinolones and terbium(III) to form luminescent chelates, which were determined at lambda(ex) 340 and lambda(em) 545 nm (FL mode) or at lambda(ex) 281 and lambda(em) 545 nm (TR mode). Dynamic ranges of the calibration graphs, obtained with standard solutions of analytes and FL and TR modes, respectively, were 190-3500 and 316-2000 ng mL-1 for marbofloxacin, 8-3500 and 8.1-1500 ng mL-1 for ciprofloxacin, 6.2-3500 and 13-1500 ng mL-1 for danofloxacin, 7.4-3500 and 8.4-1500 ng mL-1 for enrofloxacin, 14-3500 and 20-2000 ng mL-1 for sarafloxacin, 12.5-3500 and 13.9-1200 ng mL-1 for difloxacin, 7.6-3500 and 13-3000 ng mL-1 for oxolinic acid, and 9-2000 and 130-3000 ng mL-1 for flumequine. Limit of detection values obtained using FL and TR modes, respectively, were 60 and 95 ng mL-1 for marbofloxacin, 2 and 2.4 ng mL-1 for ciprofloxacin, 1.9 and 3.9 ng mL-1 for danofloxacin, 2.2 and 2.5 ng mL-1 for enrofloxacin, 3.8 and 7 ng mL-1 for sarafloxacin, 4 and 4.2 ng mL-1 for difloxacin, 2.3 and 4 ng mL-1 for oxolinic acid, and 2.7 and 40 ng mL-1 for flumequine. The precision was established at two concentration levels of each analyte and expressed as the percentage of relative standard deviation with values ranging between 1.9 and 7.8%. The validation procedure for the analysis of samples was carried out using European Community recommendations, and the decision limit and detection capability were calculated for bovine whole milk. The method was applied to whole, semiskimmed, and skimmed milk samples spiked with the target analytes, and the recoveries ranged between 93.3 and 106.0%.
authors.affiliation: Department of Analytical Chemistry, "Marie Curie Annex" Building, Campus of Rabanales, University of Córdoba, E-14071 Córdoba, Spain.
authors.forename: Rafael Carlos
authors.initials: RC
authors.lastname: Rodríguez-Díaz
authors.forename: Juan Manuel
authors.initials: JM
authors.lastname: Fernandez-Romero
authors.forename: María Paz
authors.initials: MP
authors.lastname: Aguilar-Caballos
authors.forename: Agustina
authors.initials: A
authors.lastname: Gómez-Hens
mesh_terms: D000818:Animals; D002853:Chromatography, Liquid / Q000379:methods*; D024841:Fluoroquinolones / Q000032:analysis*; D008892:Milk / Q000737:chemistry*; D011786:Quality Control; D013053:Spectrophotometry
chemical_list: D024841:Fluoroquinolones