Endovascular exclusion of popliteal artery aneurysms with expanded polytetrafluoroethylene stent-grafts: early results.
There is increasing interest in using endovascular methods instead of surgical reconstruction to treat popliteal artery aneurysms. Exclusive use of the Viabahn stent-graft, a nitinol stent covered with expanded polytetrafluoroethylene, was assessed in the treatment of patients who presented with popliteal artery aneurysms in the absence of acute limb ischemia. Technical success, endoleaks, graft patency, freedom from amputation, and aneurysm sac flow and size changes were assessed by duplex ultrasound. From June 2004 to March 2006, 16 men (mean age, 76 years; range, 65-83) underwent endovascular exclusion of 23 popliteal artery aneurysms (mean diameter, 2.5 cm; range, 1.3-6.7 cm). Nine lesions had partial thrombus on preprocedural duplex imaging. Nineteen of the 23 limbs treated had at least 2-vessel tibial artery runoff. Procedures were performed under local anesthesia using ipsilateral percutaneous antegrade arterial access. All patients received 75 mg/day of clopidogrel afterward. Follow-up assessments included direct clinical examinations and duplex ultrasonography performed 1, 3, 6, and 12 months after the procedure. Primary patency and amputation-free survival were calculated using Kaplan-Meier analysis. Complete aneurysm exclusion (technical success) was achieved in all cases. During the mean follow-up of 7 months (range, 1-21 months), 22 of 23 treated limbs remained asymptomatic. One stent-graft thrombosis occurred 6 months after the procedure and was successfully treated with percutaneous mechanical thrombectomy, balloon angioplasty of a stent-graft stenosis, and insertion of an uncovered nitinol stent. No popliteal artery aneurysm sac size enlargements or endoleaks were detected. At 12 months, the treated limb mean ankle-brachial index was 1.0 (range, 0.82-1.31) and the primary and secondary patency rates were 93% and 100%, respectively. Early results with Viabahn endovascular stent-graft exclusion of asymptomatic popliteal artery aneurysms are promising. Patient selection for endovascular repair depends on suitable popliteal artery anatomy, extent of aneurysmal degeneration, and quality of tibial arterial runoff.
Rajasinghe
Hiranya A
HA
Vascular Surgery and Endovascular Intervention, Anchor Health Centers, Naples, FL 34102, USA. raj@anchorhealthcenters.com
Tzilinis
Argyrios
A
Keller
Theresa
T
Schafer
Jeff
J
Urrea
Sandra
S
D000368:Aged; D000369:Aged, 80 and over; D000783:Aneurysm / Q000000981:diagnostic imaging / Q000503:physiopathology / Q000601:surgery*; D017130:Angioplasty; D019917:Blood Vessel Prosthesis Implantation; D005500:Follow-Up Studies; D006083:Graft Occlusion, Vascular / Q000503:physiopathology; D006801:Humans; D053208:Kaplan-Meier Estimate; D008297:Male; D018579:Patient Selection; D011138:Polytetrafluoroethylene; D011150:Popliteal Artery / Q000000981:diagnostic imaging / Q000503:physiopathology / Q000601:surgery*; D011474:Prosthesis Design; D015641:Radiography, Interventional; D012189:Retrospective Studies; D015607:Stents; D013997:Time Factors; D016896:Treatment Outcome; D014654:Vascular Patency
D011138:Polytetrafluoroethylene