title: Development of a 100 nmol mol(-1) propane-in-air SRM for automobile-exhaust testing for new low-emission requirements.
abstract: New US federal low-level automobile emission requirements, for example zero-level-emission vehicle (ZLEV), for hydrocarbons and other species, have resulted in the need by manufacturers for new certified reference materials. The new emission requirement for hydrocarbons requires the use, by automobile manufacturing testing facilities, of a 100 nmol mol(-1) propane in air gas standard. Emission-measurement instruments are required, by federal law, to be calibrated with National Institute of Standards and Technology (NIST) traceable reference materials. Because a NIST standard reference material (SRM) containing 100 nmol mol(-1) propane was not available, the US Environmental Protection Agency (EPA) and the Automobile Industry/Government Emissions Research Consortium (AIGER) requested that NIST develop such an SRM. A cylinder lot of 30 gas mixtures containing 100 nmol mol(-1) propane in air was prepared in 6-L aluminium gas cylinders by a specialty gas company and delivered to the Gas Metrology Group at NIST. Another mixture, contained in a 30-L aluminium cylinder and included in the lot, was used as a lot standard (LS). Using gas chromatography with flame-ionization detection all 30 samples were compared to the LS to obtain the average of six peak-area ratios to the LS for each sample with standard deviations of <0.31%. The average sample-to-LS ratio determinations resulted in a range of 0.9828 to 0.9888, a spread of 0.0060, which corresponds to a relative standard deviation of 0.15% of the average for all 30 samples. NIST developed its first set of five propane in air primary gravimetric standards covering a concentration range 91 to 103 nmol mol(-1) with relative uncertainties of 0.15%. This new suite of propane gravimetric standards was used to analyze and assign a concentration value to the SRM LS. On the basis of these data each SRM sample was individually certified, furnishing the desired relative expanded uncertainty of +/-0.5%. Because automobile companies use total hydrocarbons to make their measurements, it was also vital to assign a methane concentration to the SRM samples. Some of the SRM samples were analyzed and found to contain 1.2 nmol mol(-1) methane. Twenty-five of the samples were certified and released as SRM 2765.
authors.affiliation: Analytical Chemistry Division, National Institute of Standards and Technology, Gaithersburg, MD, 20899-8393, USA. george.rhoderick@nist.gov
authors.forename: George C
authors.initials: GC
authors.lastname: Rhoderick