title: Evaluation of manometric temperature measurement, a process analytical technology tool for freeze-drying: part II measurement of dry-layer resistance.
abstract: The purpose of this work was to study the factors that may cause systematic errors in the manometric temperature measurement (MTM) procedure used to determine product dry-layer resistance to vapor flow. Product temperature and dry-layer resistance were obtained using MTM software installed on a laboratory freeze-dryer. The MTM resistance values were compared with the resistance values obtained using the "vial method." The product dry-layer resistances obtained by MTM, assuming fixed temperature difference (DeltaT; 2 degrees C), were lower than the actual values, especially when the product temperatures and sublimation rates were low, but with DeltaT determined from the pressure rise data, more accurate results were obtained. MTM resistance values were generally lower than the values obtained with the vial method, particularly whenever freeze-drying was conducted under conditions that produced large variations in product temperature (ie, low shelf temperature, low chamber pressure, and without thermal shields). In an experiment designed to magnify temperature heterogeneity, MTM resistance values were much lower than the simple average of the product resistances. However, in experiments where product temperatures were homogenous, good agreement between MTM and "vial-method" resistances was obtained. The reason for the low MTM resistance problem is the fast vapor pressure rise from a few "warm" edge vials or vials with low resistance. With proper use of thermal shields, and the evaluation of DeltaT from the data, MTM resistance data are accurate. Thus, the MTM method for determining dry-layer resistance is a useful tool for freeze-drying process analytical technology.
authors.affiliation: School of Pharmacy, University of Connecticut, Storrs, CT, USA.
authors.forename: Xiaolin Charlie
authors.initials: XC
authors.lastname: Tang
authors.forename: Steven L
authors.initials: SL
authors.lastname: Nail
authors.forename: Michael J
authors.initials: MJ
authors.lastname: Pikal
mesh_terms: D003198:Computer Simulation; D005612:Freeze Drying / Q000379:methods*; D008365:Manometry / Q000379:methods*; D008422:Materials Testing / Q000379:methods; D008956:Models, Chemical; D004364:Pharmaceutical Preparations / Q000032:analysis / Q000737:chemistry*; D011312:Pressure; D012984:Software; D013678:Technology, Pharmaceutical / Q000379:methods*; D013696:Temperature; D013817:Thermography / Q000379:methods*; D014867:Water / Q000032:analysis / Q000737:chemistry
chemical_list: D004364:Pharmaceutical Preparations; D014867:Water