Results from a multicenter evaluation of the 4th generation Elecsys Troponin T assay.
OBJECTIVE
Discrepancies between serum and heparin plasma samples have been described for many commercial troponin assays including the cardiac troponin T (cTnT) assay. Using the current 3rd generation Elecsys Troponin T immunoassay, heparin plasma cannot be recommended for the determination of cTnT due to systematic lower test results caused by a direct interference of the immunoassay by heparin. The purpose of the multicenter study was to evaluate the analytical performance of an improved 4th generation Elecsys Troponin T immunoassay with a special focus on the comparability of cTnT results determined in heparin plasma and serum.


RESULTS
The multicenter evaluation was performed in 10 clinical laboratories according to a standardized protocol (Roche Diagnostics, Penzberg, Germany, Study No. B05P008). The Elecsys Troponin T immunoassay was performed on the Modular Analytics E170 and Elecsys 2010 systems. Intraassay imprecision (n = 21) and total imprecision (2 runs/d, 10 days, triplicate measurements) were evaluated using 2 commercial controls (Roche Diagnostics) and 6 different serum pools (cTnT: 0.0140 - 4.102 microg/L). Intraassay CVs ranged from 0.73 to 3.22%. Total imprecision CVs ranged from 3.61 to 35.45% (cTnT < 0.1 microg/L) and 1.82 to 9.09% (cTnT > 0.1 microg/L), respectively. The cut-off for myocardial necrosis was determined to be 0.03 microg/L using the 10% total imprecision CV criteria. Linearity was assessed by serial dilutions of 6 different serum samples using cTnT negative serum pools. Linearity was proven up to 21.3 microg/L (recoveries: 90% - 110%). Regression data of all comparison studies were calculated according to the method of Passing and Bablok. The method comparison between the 4th generation and the commercially available cTnT immunoassay showed highly similar results across the whole measuring range (0.01 - 25.0 microg/L): y = 1.024x -0.001, r = 0.998; n = 988. Using the commercially available cTnT reagent, the serum to heparin plasma comparison yielded a systematic bias to approximately 8% lower cTnT results in heparin plasma. However, suitable comparability was obtained using the 4th generation Elecsys cTnT assay. The regression analysis (serum vs. heparin plasma) across the studied measuring range (cTnT: 0.01 - 14 microg/L) yielded the following equation: y = 0.975x + 0.001; r = 0.986; n = 403. However, rare individual serum to matched heparin plasma samples still yielded poor comparability (deviation > 20%) using the 4th generation Elecsys Troponin T immunoassay.


CONCLUSIONS
Our data confirm an excellent analytical performance of the improved troponin T immunoassay. Most importantly, no systematic bias between cTnT results determined in serum and heparin plasma was observed from data obtained in 7 evaluation sites. The performance of the 4th generation Elecsys Troponin T assay is therefore comparable to other commercially available troponin immunoassays. Further studies are necessary to investigate the cause of poor comparability of cTnT results in rare individual serum to matched heparin plasma samples.
Hermsen
D
D
Institute of Clinical Chemistry and Laboratorv Diagnostics, Heinrich-Heine University, Düsseldorf, Germany. hermsen@med.uni-duesseldorf.de
Apple
F
F
Garcia-Beltràn
L
L
Jaffe
A
A
Karon
B
B
Lewandrowski
E
E
Mühlbacher
A
A
Müller
R
R
Ordóñez
J
J
Pagani
F
F
Panteghini
M
M
Plecko
T
T
Jarausch
J
J
D005069:Evaluation Studies as Topic; D006493:Heparin / Q000032:analysis / Q000097:blood; D006801:Humans; D007118:Immunoassay / Q000295:instrumentation* / Q000379:methods*; D012044:Regression Analysis; D020107:Troponin T / Q000032:analysis* / Q000097:blood
D020107:Troponin T; D006493:Heparin