title: Estimation of aerosol droplet sizes by using a modified DC-III portable droplet measurement system under laboratory and field conditions.
abstract: Modification of the DC-III portable droplet measurement system, permitting its use under field conditions, is described. Under laboratory conditions, the system effectively sampled water droplets from aerosols produced by a dry ice/water generator and high-pressure syringe. Seven droplet sizes, totaling 71,053 droplets within 22 tests (dry ice method), consisted of 1-, 2-, 6-, 11-, 18-, 25-, and 34-microm droplets with individual (rounded) percentages of 45.25, 37.22, 13.85, 3.17, 0.45, 0.02, and 0.005, respectively, for each size. Cumulatively, 1-microm droplets accounted for ca. 45.25% of the droplets sampled; combined with 2-microm (ca. 82.48% together), 6-microm (ca. 96.33% together), and 11-microm droplets, yielded ca. 99.51% of the droplets sampled. The syringe produced 12 droplet sizes, with 4,121 droplets sampled, consisting of 1, 2, 6, 11, 18, 25, 34, 45, 56, 69, 83, and 99 microm with individual percentages of 15.43, 21.91, 24.58, 17.30, 10.62, 4.65, 2.93, 1.33, 0.63, 0.33, 0.16, 0.07, respectively, for each size. The 6-microm droplets contributed the highest individual percentage, and cumulatively, these droplets combined with 1- and 2-microm droplets, yielding 61.93%, whereas 11- to 45-microm droplets contributed 36.83%, for a total of 98.76%. Droplets measuring 56-99 microm accounted for ca. 1.24% of droplets sampled. Hand-fogger oil aerosols produced 12 droplet sizes (1-38 microm) at test distances of 7.6 and 15.2 m, with 1,979 and 268 droplets sampled, respectively, during 10 tests at each distance. With analysis of variance of transformed individual percentages for each size at both distances, no significant differences were observed for 7.6 and 15.2 m. Cumulatively, 1-, 2-, 3-, and 5-microm droplets contributed 82.87 and 80.97%, whereas 8-, 11-, 14-, and 18-microm droplets added 14.55% to totals at both 7.6 and 15.2 m, respectively. Droplets measuring 22, 27, 32, and 38 microm contributed 2.57% and 4.47% to samples obtained at 7.6 and 15.2 m. The average mass median diameter (MMD) of mineral oil aerosols obtained at 7.6 and 15.2 m were 19.55 +/- 1.62 and 15.49 +/- 1.35, respectively. Positioned at 15 m (50 ft) within a field cage test plot during 12 separate tests, the MMD values obtained downrange were less than calibration MMD values in 5 tests (10-93% of calibration values), but 1.1 to 4.4 times greater in 7 remaining tests. With exception of 2 tests, susceptible Sebring strain Culex quinquefasciatus mortality at the 31-m (100-ft) stake, 15 m (50 ft) behind the DC-III, was consistently 100%. The average MMD values for malathion, permethrin, and resmethrin (at 0.003 and 0.007 rates) at this distance were 13.24 +/- 6.08, 7.48 +/- 5.69, 16.64 +/- 5.54, and 15.48 +/- 2.89, respectively.
authors.affiliation: Mosquito Control Division, Harris County Public Health & Environmental Services, Houston, TX 77021, USA.
authors.forename: James A
authors.initials: JA
authors.lastname: Dennett
authors.forename: Pamela M
authors.initials: PM
authors.lastname: Stark
authors.forename: Nathan Y
authors.initials: NY
authors.lastname: Vessey
authors.forename: Ray E
authors.initials: RE
authors.lastname: Parsons
authors.forename: Rudy
authors.initials: R
authors.lastname: Bueno
mesh_terms: D000336:Aerosols; D000392:Air Movements; D000818:Animals; D003465:Culex / Q000187:drug effects; D004367:Dry Ice; D004784:Environmental Monitoring / Q000295:instrumentation* / Q000379:methods; D007306:Insecticides / Q000008:administration & dosage* / Q000494:pharmacology; D009032:Mosquito Control / Q000295:instrumentation / Q000379:methods; D009330:Nebulizers and Vaporizers / Q000592:standards; D010316:Particle Size; D012107:Research Design / Q000592:standards; D014867:Water
chemical_list: D000336:Aerosols; D004367:Dry Ice; D007306:Insecticides; D014867:Water