title: [Computation of the cross-sectional area of the cable in the power circuit of the X-ray machine].
abstract: The source impedance of the power circuit in the x-ray machine is analyzed in the paper and based on the voltage drop generated by the impedance, the cross-sectional area of the cable is calculated. In the end, the cross-sectional areas of the cables, corresponding to their respective distances between the transformers and the switchboards are given.
authors: Sanko Medical Systems Co., Ltd. Beijing.
authors: Xin-min
authors: XM
authors: Meng
authors: Da-yu
authors: DY
authors: Feng
mesh_terms: D011211:Electric Power Supplies; D011859:Radiography / Q000295:instrumentation*