Comparing oxygen transfer performance between three membrane oxygenators: effect of temperature changes during cardiopulmonary bypass.
Recently, a new oxygenator (Dideco 903 [D903], Dideco, Mirandola, Italy) has been introduced to the perfusion community, and we set about testing its oxygen transfer performance and then comparing it to two other models. This evaluation was based on the comparison between oxygen transfer slope, gas phase arterial oxygen gradients, degree of blood shunting, maximum oxygen transfer, and diffusing capacity calculated for each membrane. Sixty patients were randomized into three groups of oxygenators (Dideco 703 [D703], Dideco; D903; and Quadrox, Jostra Medizintechnik AG, Hirrlingen, Germany) including 40/20 M/F of 68.6 +/- 11.3 years old, with a body weight of 71.5 +/- 12.1 kg, a body surface area (BSA) of 1.84 +/- 0.3 m(2), and a theoretical blood flow rate (index 2.4 times BSA) of 4.4 +/- 0.7 L/min. The maximum oxygen transfer (VO(2)) values were 313 mL O(2)/min (D703), 579 mL O(2)/min (D903), and 400 mL O(2)/min (Quadrox), with the D903 being the most superior (P < 0.05). Oxygen (O(2)) gradients were 320 mm Hg (D703), 235 mm Hg (D903), and 247 mm Hg (Quadrox), meaning D903 and Quadrox are more efficient versus the D703 (P < 0.05). Shunt fraction (Qs/Qt) and diffusing capacity (DmO(2)) were comparable (P = ns). Diffusing capacity values indexed to BSA (DmO(2)/m(2)) were 0.15 mL O(2)/min/mm Hg/m(2) (D703), 0.2 mL O(2)/min/mm Hg/m(2) (D903), and 0.18 mL O(2)/min/mm Hg/m(2) (Quadrox) with D903 outperforming D703 (P < 0.0005). During hypothermia (32.0 +/- 0.3 degrees C), there was a lower absolute and relative VO(2 )for all three oxygenators (P = ns). The O(2) gradients, DmO(2) and DmO(2)/m(2), were significantly lower for all oxygenators (P < 0.01). Also, Qs/Qt significantly rose for all oxygenators (P < 0.01). The oxygen transfer curve is characteristic to each oxygenator type and represents a tool to quantify oxygenator performance. Using this parameter, we demonstrated significant differences among commercially available oxygenators. However, all three oxygenators are considered to meet the oxygen needs of the patients.
Jegger
David
D
Department of Cardiovascular Surgery, Centre Hospitalier Universitaire Vaudois (CHUV), Lausanne, Switzerland. David.Jegger@chuv.ch
Tevaearai
Hendrik T
HT
Mallabiabarrena
Iker
I
Horisberger
Judith
J
Seigneul
Isabelle
I
von Segesser
Ludwig K
LK
D000368:Aged; D002315:Cardiopulmonary Bypass / Q000295:instrumentation* / Q000379:methods; D001026:Coronary Artery Bypass / Q000295:instrumentation* / Q000379:methods; D015199:Extracorporeal Membrane Oxygenation / Q000295:instrumentation* / Q000379:methods; D005260:Female; D006801:Humans; D007036:Hypothermia, Induced; D008297:Male; D010100:Oxygen / Q000378:metabolism*; D010101:Oxygen Consumption; D010107:Oxygenators, Membrane; D013696:Temperature
D010100:Oxygen