High-resolution magnetic resonance angiography of the lower extremities with a dedicated 36-element matrix coil at 3 Tesla.
BACKGROUND
Recent developments in hard- and software help to significantly increase image quality of magnetic resonance angiography (MRA). Parallel acquisition techniques (PAT) help to increase spatial resolution and to decrease acquisition time but also suffer from a decrease in signal-to-noise ratio (SNR). The movement to higher field strength and the use of dedicated angiography coils can further increase spatial resolution while decreasing acquisition times at the same SNR as it is known from contemporary exams. The goal of our study was to compare the image quality of MRA datasets acquired with a standard matrix coil in comparison to MRA datasets acquired with a dedicated peripheral angio matrix coil and higher factors of parallel imaging.


METHODS
Before the first volunteer examination, unaccelerated phantom measurements were performed with the different coils. After institutional review board approval, 15 healthy volunteers underwent MRA of the lower extremity on a 32 channel 3.0 Tesla MR System. In 5 of them MRA of the calves was performed with a PAT acceleration factor of 2 and a standard body-matrix surface coil placed at the legs. Ten volunteers underwent MRA of the calves with a dedicated 36-element angiography matrix coil: 5 with a PAT acceleration of 3 and 5 with a PAT acceleration factor of 4, respectively. The acquired volume and acquisition time was approximately the same in all examinations, only the spatial resolution was increased with the acceleration factor. The acquisition time per voxel was calculated. Image quality was rated independently by 2 readers in terms of vessel conspicuity, venous overlay, and occurrence of artifacts. The inter-reader agreement was calculated by the kappa-statistics. SNR and contrast-to-noise ratios from the different examinations were evaluated.


RESULTS
All 15 volunteers completed the examination, no adverse events occurred. None of the examinations showed venous overlay; 70% of the examinations showed an excellent vessel conspicuity, whereas in 50% of the examinations artifacts occurred. All of these artifacts were judged as none disturbing. Inter-reader agreement was good with kappa values ranging between 0.65 and 0.74. SNR and contrast-to-noise ratios did not show significant differences.


CONCLUSIONS
Implementation of a dedicated coil for peripheral MRA at 3.0 Tesla helps to increase spatial resolution and to decrease acquisition time while the image quality could be kept equal. Venous overlay can be effectively avoided despite the use of high-resolution scans.
Kramer
Harald
H
Institute for Clinical Radiology, University Hospitals of Munich-Grosshadern Campus, Ludwig-Maximilians-University, Munich, Germany. harald.kramer@med.uni-muenchen.de
Michaely
Henrik J
HJ
Matschl
Volker
V
Schmitt
Peter
P
Reiser
Maximilian F
MF
Schoenberg
Stefan O
SO
D000328:Adult; D003287:Contrast Media / Q000008:administration & dosage; D004867:Equipment Design; D019786:Gadolinium DTPA / Q000008:administration & dosage; D006801:Humans; D007091:Image Processing, Computer-Assisted; D007866:Leg / Q000098:blood supply*; D018810:Magnetic Resonance Angiography / Q000295:instrumentation / Q000379:methods*; D008297:Male; D008875:Middle Aged; D019047:Phantoms, Imaging
D003287:Contrast Media; D019786:Gadolinium DTPA