4T split TEM volume head and knee coils for improved sensitivity and patient accessibility.
Split RF coils offer improved patient access by eliminating the need for the coil to be slid over the region of interest. For unshielded birdcage coils, the presence of end ring currents necessitates a direct electrical connection between two halves of the coil. For high-field (>3T) shielded birdcage coils, both the shield and the coil must be split and reliably connected electrically. This problem can be circumvented by the use of split TEM volume coils. Since the elements of a TEM coil are coupled inductively, no direct electrical connection between the halves is necessary. In this work we demonstrate that the effects of splitting the shield for head and knee TEMs can be compensated for, and performance retained. For the knee, the improved access allowed the coil diameter to be reduced, enhancing the sensitivity by 15-20%.
Avdievich
Nikolai I
NI
Department of Radiology, Albert Einstein College of Medicine, Bronx, NY, USA. nikolai.avdievich@yale.edu
Bradshaw
Ken
K
Lee
Jing-Huei
JH
Kuznetsov
Andrey M
AM
Hetherington
Hoby P
HP
D004867:Equipment Design; D006257:Head; D006801:Humans; D007717:Knee; D008279:Magnetic Resonance Imaging / Q000295:instrumentation*; D011846:Radio Waves; D012680:Sensitivity and Specificity