Dosin--a new pump device. Its construction and function.
The cyclic flow of the Dosin is useful in interventional radiology since it prevents clotting and increases the mixing with the blood. The flow rate varies from 0.03 mL/s to 5 mL/s and it is recommended to use fluid with a viscosity lower than 3 mPa.s. The outlet pressure is recommended below 300 mm Hg. The Dosin must be replaced after each patient, alternatively after maximum 24h continuous use.
Wahlberg
Andreas
A
Department of Diagnostic Radiology, Akademiska Sjukhuset, Uppsala University, Sweden.
Erikson
Uno
U
Fuchs
Laszlo
L
Andersson
Anders
A
Attefall
Anna
A
D004867:Equipment Design; D006801:Humans; D007260:Infusion Pumps; D015642:Radiology, Interventional