Penalized maximum-likelihood sinogram restoration for dual focal spot computed tomography.
Due to various system non-idealities, the raw data generated by a computed tomography (CT) machine are not readily usable for reconstruction. Although the deterministic nature of corruption effects such as crosstalk and afterglow permits correction by deconvolution, there is a drawback because deconvolution usually amplifies noise. Methods that perform raw data correction combined with noise suppression are commonly termed sinogram restoration methods. The need for sinogram restoration arises, for example, when photon counts are low and non-statistical reconstruction algorithms such as filtered backprojection are used. Many modern CT machines offer a dual focal spot (DFS) mode, which serves the goal of increased radial sampling by alternating the focal spot between two positions on the anode plate during the scan. Although the focal spot mode does not play a role with respect to how the data are affected by the above-mentioned corruption effects, it needs to be taken into account if regularized sinogram restoration is to be applied to the data. This work points out the subtle difference in processing that sinogram restoration for DFS requires, how it is correctly employed within the penalized maximum-likelihood sinogram restoration algorithm and what impact it has on image quality.
Forthmann
P
P
Philips Research Europe - Hamburg, Germany. peter.forthmann@philips.com
Köhler
T
T
Begemann
P G C
PG
Defrise
M
M
D000465:Algorithms; D003627:Data Interpretation, Statistical; D016013:Likelihood Functions; D011856:Radiographic Image Enhancement / Q000379:methods; D011857:Radiographic Image Interpretation, Computer-Assisted / Q000379:methods*; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity; D014057:Tomography, X-Ray Computed / Q000379:methods*