Why do nickel-titanium archwires fracture intraorally? Fractographic analysis and failure mechanism of in-vivo fractured wires.
BACKGROUND
The aim of this study was to characterize intraorally fractured nickel-titanium (Ni-Ti) archwires, determine the type of fracture, assess changes in the alloy's hardness and structure, and propose a mechanism of failure.


METHODS
Eleven Ni-Ti SE 200 and 19 copper-Ni-Ti (both, Ormco, Glendora, Calif) intraorally fractured archwires were collected. The location of fracture (anterior or posterior), wire type, cross section, and period of service before fracture were recorded. The retrieved wires and brand-, type-, and size-matched specimens of unused wires were subjected to scanning electron microscopy to assess the fracture type and morphological variation of fracture site of retrieved specimens, and to Vickers hardness (HV200) testing to investigate the hardness of as-received and in-vivo fractured specimens. Fracture site distribution was statistically analyzed with the chi-square test (alpha = 0.05), whereas the results of the hardness testing were analyzed with 2-way ANOVA with state (control vs in-vivo fractured) and composition (Ni-Ti SE vs copper-Ni-Ti) serving as discriminating variables and the Student-Newman-Keuls test at the 95% confidence level.


RESULTS
The fracture site distribution showed a preferential location at the midspan between the premolar and the molar, suggesting that masticatory forces and complex loading during engagement of the wire to the bracket slot and potential intraoral aging might account for fracture incidence. All retrieved wires had the distinct features of brittle fracture without plastic deformation or crack propagation, whereas no increase in hardness was observed for the retrieved specimens.


CONCLUSIONS
Most fractures sites were in the posterior region of the arch, probably because of the high-magnitude masticatory forces. Brittle fracture without plastic deformation was observed in most Ni-Ti wires regardless of archwire composition. There was no increase in the hardness of the intraorally exposed specimens regardless of wire type. This contradicts previous in-vitro studies and rules out hydrogen embrittlement as the cause of fracture.
Zinelis
Spiros
S
Department of Biomaterials, School of Dentistry, University of Athens, Athens, Greece.
Eliades
Theodore
T
Pandis
Nikolaos
N
Eliades
George
G
Bourauel
Christoph
C
D000149:Acidulated Phosphate Fluoride / Q000737:chemistry; D000704:Analysis of Variance; D003300:Copper; D003343:Corrosion; D003722:Dental Alloys; D019544:Equipment Failure Analysis; D006244:Hardness; D006801:Humans; D006859:Hydrogen / Q000737:chemistry; D009532:Nickel; D009969:Orthodontic Wires; D013718:Tensile Strength; D014025:Titanium
C415948:Copper NiTi; D003722:Dental Alloys; C040654:titanium nickelide; D000149:Acidulated Phosphate Fluoride; D003300:Copper; D009532:Nickel; D006859:Hydrogen; D014025:Titanium