Circumferential lesion formation around the pulmonary veins in the left atrium with focused ultrasound using a 2D-array endoesophageal device: a numerical study.
Atrial fibrillation (AF) is the most frequently sustained cardiac arrhythmia affecting humans. The electrical isolation by ablation of the pulmonary veins (PVs) in the left atrium (LA) of the heart has been proven as an effective cure of AF. The ablation consists mainly in the formation of a localized circumferential thermal coagulation of the cardiac tissue surrounding the PVs. In the present numerical study, the feasibility of producing the required circumferential lesion with an endoesophageal ultrasound probe is investigated. The probe operates at 1 MHz and consists of a 2D array with enough elements (114 x 20) to steer the acoustic field electronically in a volume comparable to the LA. Realistic anatomical conditions of the thorax were considered from the segmentation of histological images of the thorax. The cardiac muscle and the blood-filled cavities in the heart were identified and considered in the sound propagation and thermal models. The influence of different conditions of the thermal sinking in the LA chamber was also studied. The circumferential ablation of the PVs was achieved by the sum of individual lesions induced with the proposed device. Different scenarios of lesion formation were considered where ultrasound exposures (1, 2, 5 and 10 s) were combined with maximal peak temperatures (60, 70 and 80 degrees C). The results of this numerical study allowed identifying the limits and best conditions for controlled lesion formation in the LA using the proposed device. A controlled situation for the lesion formation surrounding the PVs was obtained when the targets were located within a distance from the device in the range of 26 +/- 7 mm. When combined with a maximal temperature of 70 degrees C and an exposure time between 5 and 10 s, this distance ensured preservation of the esophageal structures, controlled lesion formation and delivery of an acoustic intensity at the transducer surface that is compatible with existing materials. With a peak temperature of 70 degrees C, the device and setup presented here induced highly localized lesions with a lesion volume varying from 10 +/- 4 to 18 +/- 7 mm(3) for an ultrasound exposure between 5 and 10 s, respectively, while the intensity varied from 26 +/- 7 to 20 +/- 6 W cm(-2).
Pichardo
Samuel
S
Imaging Research-Sunnybrook Health Sciences Centre, 2075 Bayview Avenue, Room C713, Toronto, Ontario, Canada. samuel.pichardo@sri.utoronto.ca
Hynynen
Kullervo
K
D001281:Atrial Fibrillation / Q000473:pathology / Q000503:physiopathology* / Q000601:surgery*; D003198:Computer Simulation; D004867:Equipment Design; D019544:Equipment Failure Analysis; D004947:Esophagus; D006329:Heart Conduction System / Q000473:pathology / Q000503:physiopathology* / Q000601:surgery; D006801:Humans; D008954:Models, Biological; D011667:Pulmonary Veins / Q000473:pathology / Q000503:physiopathology* / Q000601:surgery*; D013813:Therapy, Computer-Assisted / Q000379:methods; D014159:Transducers; D014464:Ultrasonic Therapy / Q000295:instrumentation* / Q000379:methods