A small change in the design of a slit bioaerosol impactor significantly improves its collection characteristics.
While several methods are available for bioaerosol monitoring, impaction remains the most common one, particularly for collecting fungal spores. Earlier studies have shown that the collection efficiency of many conventional single-stage bioaerosol impactors falls below 50% for spores with an aerodynamic diameter between 1.7 and 2.5 microm because their cut-off size is 2.5 microm or greater. The cut-off size reduction is primarily done by substantially increasing the sampling flow rate or decreasing the impaction jet size, W, to a fraction of a millimetre, with both measures often impractical to implement. Some success has recently been reported on the utilization of an ultra-low jet-to-plate distance, S (S/W < 0.1), in circular impactors. This paper describes a laboratory evaluation and some field testing of two single-stage, single-nozzle, slit bioaerosol impactors, Allergenco-D and Air-O-Cell, which feature the same jet dimensions and flow rate but have some design configuration differences that were initially thought to be of low significance. The collection efficiency and the spore deposit characteristics were determined in the laboratory using real-time aerosol spectrometry and different microscopic enumeration methods as the test impactors were challenged with the non-biological polydisperse NaCl aerosol and the aerosolized fungal spores of Cladosporium cladosporioides, Aspergillus versicolor, and Penicillium melinii. The tests showed that a relatively small reduction in the jet-to-plate distance of a single-stage, single-nozzle impactor with a tapered inlet nozzle, combined with adding a straight section of sufficient length, can significantly decrease the cut-off size to the level that is sufficient to efficiently collect spores of all fungal species. Furthermore, it appears that the slit jet design may improve the application of partial spore counting methodologies with respect to those applied to circular deposits. Data from a demonstration field study, conducted with the two samplers in environments containing a variety of fungal species, supported the laboratory findings.
Grinshpun
Sergey A
SA
Center for Health-Related Aerosol Studies, Department of Environmental Health, University of Cincinnati, Cincinnati, Ohio 45267-0056, USA. Sergey.Grinshpun@uc.edu
Adhikari
Atin
A
Cho
Seung-Hyun
SH
Kim
Ki-Youn
KY
Lee
Taekhee
T
Reponen
Tiina
T
D000336:Aerosols; D000391:Air Microbiology; D001230:Aspergillus / Q000302:isolation & purification; D002963:Cladosporium / Q000302:isolation & purification; D015169:Colony Count, Microbial; D004784:Environmental Monitoring / Q000295:instrumentation* / Q000379:methods*; D004867:Equipment Design; D010316:Particle Size; D010407:Penicillium / Q000302:isolation & purification; D013172:Spores, Fungal / Q000302:isolation & purification*
D000336:Aerosols