Membraneless vaporization unit for direct analysis of solid sample.
A new apparatus, called 'membraneless vaporization' (MBL-VP) unit was designed and developed for direct analysis of solid samples. Solid analyte was converted into a gaseous form which then reacts with an indicator reagent. Change in absorbance was used to quantitate the analyte. Stirring with a magnetic bar was employed to facilitate the evaporation of the gas. Unlike the pervaporation technique, hydrophobic membrane was not required for this MBL-VP technique. Application of the membraneless technique for direct determination of calcium carbonate in calcium supplements, has shown to be very precise (R.S.D.=2.5% for 0.16 mmol CO3(2-)), with detection limit of 0.5 mg CaCO3. Results by this method agreed well with flame atomic absorption spectrometric method. Sample throughput was 20 samples h(-1).
Sereenonchai
K
K
Flow Innovation-Research for Science and Technology Laboratories (First Labs), Department of Chemistry, Faculty of Science, Mahidol University, Rama 6 Road, Bangkok 10400, Thailand.
Saetear
P
P
Amornthammarong
N
N
Uraisin
K
K
Wilairat
P
P
Motomizu
S
S
Nacapricha
D
D