[The estimated glomerular filtration rate in outpatients].
The glomerular filtration rate (GFR) calculated by the MDRD formula was estimated in a population of outpatients aged over 18 years. Serum creatinine concentrations were measured, by using 5 analytical systems: Abbott Architect (n = 9054), Roche Modular (n = 22947), Roche Integra 400 (n = 2748), Roche Integra 700 (n = 8350), and Roche Hitachi (n = 20196). For 4 systems, the distribution of GFR differed little. The exception was the Roche Hitachi analytical system where TE and bias were higher than the acceptable requirements. Therefore, with this system, the proportion of outpatients with a GFR of < 60 ml/min per 1.73 m2 was twice higher than that with the use other analytical systems. Such publications should contain data on the analytical quality of measurement of serum creatinine concentrations in the range of 88 to 140 micromol/l. The estimated GFR may be clinically used provided that the performance of this test will be better than the acceptable analytical quality requirements established by the working NKDEP group.
Moshkin
A V
AV
Aref'eva
I A
IA
Sveshchinskaia
I I
II
Kondrasheva
E A
EA
D000293:Adolescent; D000328:Adult; D000368:Aged; D000369:Aged, 80 and over; D002908:Chronic Disease; D003404:Creatinine / Q000097:blood*; D005260:Female; D005919:Glomerular Filtration Rate; D006801:Humans; D007674:Kidney Diseases / Q000175:diagnosis / Q000503:physiopathology; D008297:Male; D008657:Metabolic Clearance Rate; D008875:Middle Aged; D010045:Outpatients; D012189:Retrospective Studies
D003404:Creatinine