An in vitro evaluation of the accuracy of Dentaport ZX apex locator in enlarged root canals.
BACKGROUND
The aim of this study was to examine the effects of root canal irrigants on the accuracy of Dentaport ZX electronic apex locator (EAL) in enlarged root canals. Our previous study revealed that as the diameter of the root canal increased, the electronically measured length with small size files became shorter in the presence of blood. It is not known whether different canal irrigants would interfere with the reading accuracy of an EAL in enlarged root canals.


METHODS
A total of 45 extracted single-rooted human teeth were used. In Stage 1, canals were instrumented using #10-40 K-files with a #40 K-file as the master apical file (MAF). The teeth were randomly divided into five groups and mounted in an experimental apparatus. The following irrigants were used during electronic canal measurements: Group A = 0.5% NaOCl; Group B = 2.5% NaOCl; Group C = 15% EDTA; Group D = 0.8% chlorhexidine (CHX); and Group E = RC Prep. In Stage 2, the canals were enlarged using a #60 K-file as the MAF. In Stages 1 and 2, the apical portion of the canals was instrumented using the step-back sequence (up to a #80 K-file). In Stage 3, the canals were enlarged again using a #80 K-file as the MAF. In each stage, the canal length was measured with a Dentaport ZX using #10 and #40, #10 and #60, and #10 and #80 K-files for Stages 1, 2 and 3, respectively. Data were analysed by two-way ANOVA and Fisher's PLSD test.


RESULTS
Statistical analysis showed significant differences among all groups (p < 0.01). There was no significant difference between Group A and B at each stage and between Group D and E at Stage 2. A statistical significant difference was observed between #10 K-file and other files (#40, #60 and #80) at each stage (p < 0.01). In Groups A and B, there was no significant difference between #10 and #40 K-files (Stage 1) and between #10 and #60 K-files (Stage 2).


CONCLUSIONS
The Dentaport ZX was accurate and not adversely affected by the presence of 0.5% or 2.5% NaOCl and EDTA in the enlarged canals, and the measured lengths obtained with small and large size files were comparable. However, it was accurate in the presence of CHX and RC Prep only when large size files were used, and the length measured with small size files was greater than the actual length.
Ebrahim
A K
AK
Pulp Biology and Endodontics, Department of Restorative Sciences, Graduate School, Tokyo Medical and Dental University, Japan. aqeel_endo@hotmail.com
Wadachi
R
R
Suda
H
H
D000704:Analysis of Variance; D003762:Dental Instruments; D003786:Dental Pulp Cavity / Q000033:anatomy & histology; D006801:Humans; D009811:Odontometry / Q000295:instrumentation*; D012388:Root Canal Irrigants; D018915:Root Canal Preparation / Q000295:instrumentation*; D019227:Tooth Apex / Q000033:anatomy & histology*
D012388:Root Canal Irrigants