Control aspects of motor neural prosthesis: sensory interface.
A neural prosthesis (NP) has two applications: permanent assistance of function, and temporary assistance that contributes to long-term recovery of function. Here, we address control issues for a therapeutic NP which uses surface electrodes. We suggest that the effective NP for therapy needs to implement rule-based control. Rule-based control relies on the triggering of preprogrammed sequences of electrical stimulation by the sensory signals. The sensory system in the therapeutic NP needs to be simple for installation, allow self-calibration, it must be robust, and sufficiently redundant in order to guarantee safe operation. The sensory signals need to generate control signals; hence, sensory fusion is needed. MEMS technology today provides sensors that fulfill the technical requirements (accelerometers, gyroscopes, force sensing resistors). Therefore, the task was to design a sensory signal processing method from the mentioned solid state sensors that would recognize phases during the gait cycle. This is necessary for the control of multi channel electrical stimulation. The sensory fusion consists of the following two phases: 1) estimation of vertical and horizontal components of the ground reaction force, center of pressure, and joint angles from the solid-state sensors, and 2) fusion of the estimated signals into a sequence of command signals. The first phase was realized by the use of artificial neural networks and adaptive neuro-fuzzy inference systems, while the second by the use of inductive learning described in our earlier work [1].
Popović
Dejan B
DB
SMI, Aalborg University, Denmark, dbp@smi.auc.dk
Dosen
Strahinja
S
Popović
Mirjana B
MB
Stefanović
Filip
F
Kojović
Jovana
J
D000328:Adult; D001186:Artificial Limbs; D005260:Female; D005684:Gait / Q000502:physiology*; D006801:Humans; D008297:Male; D016571:Neural Networks, Computer; D011474:Prosthesis Design; D012815:Signal Processing, Computer-Assisted