Improvement of hemolysis in a centrifugal blood pump with hydrodynamic bearings and semi-open impeller.
We have developed a centrifugal blood pump with hydrodynamic bearings and semi-open impeller, and evaluated the levitation performance test and the hemolysis test. This pump is operated without any complicated control circuit and displacement-sensing module. The casing diameter is 74 mm and the height is 38 mm including flanges for volts. The weight is 251 g and the volume is 159 cm3. By changing the stator relative position against the rotor, the levitation characteristics of the impeller can be adjusted. The diameter of impeller is 36 mm and the height is 25 mm. The impeller is levitated by the thrust bearing of spiral groove type and a radial bearing of herringbone type. The pump performance was evaluated through the levitation performance test and the hemolysis test. As a result, the normalized index of hemolysis (NIH) was reduced from 0.72 g/100 L to 0.024 g/100 L corresponding to the changes of the groove direction of the hydrodynamic bearing and the expansion of the bearing gap. During these studies, we confirmed that the hemolytic property was improved by balancing the fluid dynamic force and the magnetic force.
Kosaka
Ryo
R
National Institute of Advanced Industrial Science and Technology (AIST), Tsukuba, Japan. ryo.kosaka@aist.go.jp
Yamane
Takashi
T
Maruyama
Osamu
O
Nishida
Masahiro
M
Yada
Toru
T
Saito
Sakae
S
Hirai
Shusaku
S
D000818:Animals; D002417:Cattle; D006461:Hemolysis; D006801:Humans; D007260:Infusion Pumps / Q000009:adverse effects; D033081:Shear Strength; D013314:Stress, Mechanical