Correlation between low triggered electromyographic thresholds and lumbar pedicle screw malposition: analysis of 4857 screws.
METHODS
A retrospective analysis of 1078 spinal surgical procedures with lumbar pedicle screw placement at a single institution.


OBJECTIVE
Based on previously established normative values, triggered electromyographic stimulation (TrgEMG) was re-examined to evaluate its efficacy in determining screw malposition.


BACKGROUND
Threshold values for confirmation of intraosseous placement of pedicle screws with EMG stimulation is controversial.


METHODS
TrgEMG threshold values for 4857 pedicle screws placed from L2 to S1 from 1996 to 2005 were analyzed. An ascending method of constant current stimulation was applied to each pedicle screw to obtain a compound muscle action potential (CMAP) from lower extremity myotomes. Previously determined threshold value normative data from a published clinical series of 233 screws were as follows: 0 to 4 mA, high likelihood of pedicle wall breach; 4 to 8 mA, possible pedicle wall breach; >8 mA, no pedicle wall defect.


RESULTS
A total of 7.74% (376 of 4857) of all screws tested had threshold values <8.0 mA. A total of 19.1% (72 of 376) of these were <4.0 mA: 54% (39 of 72) were repositioned (26) or removed (13) while the remaining 33 screws were left in place following repalpation. A total of 80.9% (304 of 376) had thresholds between 4 and 8 mA: 17.4% (53) were repositioned (38) or removed (15). Nine screws had thresholds of <or=2.8 mA and were either repositioned or removed following confirmation of a medial wall breach. A total of 74.5% (280 of 376) of all screws with thresholds <8.0 mA were verified as correctly placed by repalpation/radiography and therefore left in place.


CONCLUSIONS
The probability of a medial breach pedicle screw detected by triggered EMG stimulation increases with decreasing triggered EMG thresholds: 0.31% for >8.0 mA, 17.4% for 4.0 to 8.0 mA, 54.2% for <4.0 mA, and 100% for <2.8 mA. At 2.8 mA, triggered EMG has a specificity of 100%, with sensitivity of 8.4%; at 4.0 mA, specificity of 99% and sensitivity of 36%; and at 8.0 mA, 94% specificity and 86% sensitivity. TrgEMG is an adjunct technique and should always be used in conjunction with palpation and radiography to optimize safe pedicle screw placement.
Raynor
Barry L
BL
Washington University Medical Center, Department of Orthopaedic Surgery, Barnes-Jewish Hospital Health Systems, St. Louis, MO 63110, USA.
Lenke
Lawrence G
LG
Bridwell
Keith H
KH
Taylor
Brett A
BA
Padberg
Anne M
AM
D000200:Action Potentials; D000293:Adolescent; D000328:Adult; D000368:Aged; D000369:Aged, 80 and over; D001499:Bayes Theorem; D001863:Bone Screws; D002648:Child; D004576:Electromyography / Q000592:standards*; D005188:False Negative Reactions; D005189:False Positive Reactions; D005260:Female; D006801:Humans; D008159:Lumbar Vertebrae / Q000601:surgery*; D008297:Male; D008875:Middle Aged; D016343:Monitoring, Intraoperative / Q000592:standards*; D018482:Muscle, Skeletal / Q000294:innervation; D015203:Reproducibility of Results; D012189:Retrospective Studies; D012447:Sacrum / Q000601:surgery*; D013123:Spinal Fusion / Q000295:instrumentation*