A 3He counter version of the Thermo Fisher Scientific NRD neutron rem meter.
Thermo Fisher Scientific's NRD rem meter has been in production for almost 40 y and is the primary rem meter in use at many U.S. Department of Energy facilities. An upgrade project was initiated at the Los Alamos National Laboratory with the primary goal of increasing the NRD's neutron sensitivity through the substitution of pressurized 3He gas (4 atmospheres) for the stock counter tube's BF3 fill gas. Historically, BF3 counters were far less expensive relative to 3He and were usually chosen on the basis of cost. That is no longer the case, with pricing for both types of counters being similar. Test results have shown that the 3He counter version of the NRD exhibits stable operation at a reasonable bias voltage and good gamma rejection. Sensitivity has been increased by about a factor of four with no penalty in cost.
Olsher
Richard H
RH
Health Physics Measurements Group, Mail Stop J573, Los Alamos National Laboratory, PO Box 1663, Los Alamos, NM 87545, USA. dick@lanl.gov
Seagraves
David T
DT
D006371:Helium / Q000737:chemistry; D007554:Isotopes; D009010:Monte Carlo Method; D009502:Neutrons; D011829:Radiation Dosage; D011874:Radiometry / Q000295:instrumentation*
D007554:Isotopes; D006371:Helium