In-vitro evaluation of coronary stents and 64-detector-row computed tomography using a newly developed model of coronary artery stenosis.
BACKGROUND
Stent implantation is the predominant therapy for non-surgical myocardial revascularization in patients with coronary artery disease. However, despite substantial advances in multidetector computed tomography (MDCT) coronary imaging, a reliable detection of coronary in-stent restenosis is currently not possible.


OBJECTIVE
To examine the ability of 64-detector-row CT to detect and to grade in-stent stenosis in coronary stents using a newly developed ex-vivo vessel phantom with a realistic CT density pattern, artificial stenosis, and a thorax phantom.


METHODS
Four different stents (Liberté and Lunar ROX, Boston Scientific; Driver, Medtronic; Multi-Link Vision, Guidant) were examined. The stents were placed on a polymer tube with a diameter of 2.5, 3.0, 3.5, or 4.0 mm. Different degrees of stenosis (0%, 30%, 50%, 70-80%) were created inside the tube. For quantitative analysis, attenuation values were measured in the non-stenotic vessel outside the stent, in the non-stenotic vessel inside the stent, and in the stenotic area inside the stent. The grade of stenosis was visually assessed by two observers.


RESULTS
All stents led to artificial reduction of attenuation, the least degree of which was found in the Liberté stent (11.3+/-10.2 HU) and the Multi-Link Vision stent (17.6+/-17.9 HU; P = 0.25). Overall, the non-stenotic vessel was correctly diagnosed in 55.5%, the low-grade stenosis in 58.3%, the intermediate stenosis in 63.8%, and the high-grade stenosis in 80.5%. In the 3.0-, 3.5-, and 4.0-mm vessels, in none of the cases was a non-stenotic or low-grade stenotic vessel misdiagnosed as intermediate or high-grade stenosis. The average deviation from the real grade of stenosis was 0.40 for the Liberté stent, 0.46 for the Lunar ROX stent, 0.45 for the Driver stent, and 0.58 for the Multi-Link Vision stent.


CONCLUSIONS
Our ex-vivo data show that non-stenotic stents and low-grade in-stent stenosis can be reliably differentiated from intermediate and high-grade in-stent stenosis in vessels with a diameter of 3 to 4 mm. With regard to artifacts and the grading of stenoses, the Liberté stent was best suited for CT coronary angiography.
Schlosser
T
T
Cardiovascular Center Bethanien (CCB), Frankfurt, Germany. thomas.schlosser@uni-duisburg-essen.de
Scheuermann
T
T
Ulzheimer
S
S
Mohrs
O K
OK
Kühling
M
M
Albrecht
P E
PE
Voigtländer
T
T
Barkhausen
J
J
Schmermund
A
A
D017023:Coronary Angiography / Q000295:instrumentation* / Q000379:methods; D023903:Coronary Restenosis / Q000175:diagnosis*; D023921:Coronary Stenosis / Q000175:diagnosis*; D006801:Humans; D008954:Models, Biological; D015588:Observer Variation; D019047:Phantoms, Imaging; D015203:Reproducibility of Results; D012720:Severity of Illness Index; D015607:Stents; D014057:Tomography, X-Ray Computed / Q000295:instrumentation* / Q000379:methods