Enhancement of SPES source performances.
Installation of SPES source at LNL was finished in July 2006 and the first beam was extracted in September 2006. Commissioning results confirmed very good performance of the extracted current density. Conversely, source reliability was very poor due to glow-discharge phenomena, which were caused by the ion source axial magnetic field protruding in the high-voltage column. This problem was fixed by changing the stainless steel plasma electrode support with a ferromagnetic one. This new configuration required us to recalculate ion source solenoids positions and fields in order to recover the correct resonance pattern. Details on magnetic simulations and experimental results of high voltage column shielding are presented.
Fagotti
E
E
Consorzio RFX, Corso Stati Uniti, Padova, Italy.
Palmieri
A
A
Ren
X
X