Sampling, storage, and analysis of C2-C7 non-methane hydrocarbons from the US National Oceanic and Atmospheric Administration Cooperative Air Sampling Network glass flasks.
An analytical technique was developed to analyze light non-methane hydrocarbons (NMHC), including ethane, propane, iso-butane, n-butane, iso-pentane, n-pentane, n-hexane, isoprene, benzene and toluene from whole air samples collected in 2.5l-glass flasks used by the National Oceanic and Atmospheric Administration, Earth System Research Laboratory, Global Monitoring Division (NOAA ESRL GMD, Boulder, CO, USA) Cooperative Air Sampling Network. This method relies on utilizing the remaining air in these flasks (which is at below-ambient pressure at this stage) after the completion of all routine greenhouse gas measurements from these samples. NMHC in sample aliquots extracted from the flasks were preconcentrated with a custom-made, cryogen-free inlet system and analyzed by gas chromatography (GC) with flame ionization detection (FID). C2-C7 NMHC, depending on their ambient air mixing ratios, could be measured with accuracy and repeatability errors of generally < or =10-20%. Larger deviations were found for ethene and propene. Hexane was systematically overestimated due to a chromatographic co-elution problem. Saturated NMHC showed less than 5% changes in their mixing ratios in glass flask samples that were stored for up to 1 year. In the same experiment ethene and propene increased at approximately 30% yr(-1). A series of blank experiments showed negligible contamination from the sampling process and from storage (<10 pptv yr(-1)) of samples in these glass flasks. Results from flask NMHC analyses were compared to in-situ NMHC measurements at the Global Atmospheric Watch station in Hohenpeissenberg, Germany. This 9-months side-by-side comparison showed good agreement between both methods. More than 94% of all data comparisons for C2-C5 alkanes, isoprene, benzene and toluene fell within the combined accuracy and precision objectives of the World Meteorological Organization Global Atmosphere Watch (WMO-GAW) for NMHC measurements.
Pollmann
Jan
J
Institute of Arctic and Alpine Research (INSTAAR), 1560 30th Street, Boulder, CO 80305, USA.
Helmig
Detlev
D
Hueber
Jacques
J
Plass-Dülmer
Christian
C
Tans
Pieter
P
D000388:Air / Q000032:analysis*; D053000:Analytic Sample Preparation Methods / Q000379:methods*; D002138:Calibration; D002849:Chromatography, Gas; D005410:Flame Ionization; D005740:Gases / Q000737:chemistry; D005898:Glass / Q000737:chemistry*; D006813:Humidity; D006838:Hydrocarbons / Q000032:analysis* / Q000737:chemistry* / Q000302:isolation & purification; D008697:Methane; D010126:Ozone; D012015:Reference Standards; D012997:Solvents / Q000737:chemistry; D014481:United States; D037041:United States Government Agencies
D005740:Gases; D006838:Hydrocarbons; D012997:Solvents; D010126:Ozone; D008697:Methane