Validation of a multi-frequency electrical impedance tomography (mfEIT) system KHU Mark1: impedance spectroscopy and time-difference imaging.
Validation and interpretation of reconstructed images using a multi-frequency electrical impedance tomography (mfEIT) requires a conductivity phantom including imaging objects with known complex conductivity (sigma + iomegaepsilon) spectra. We describe imaging experiments using the recently developed mfEIT system called the KHU Mark1 with the frequency range of 10 Hz to 500 kHz. Using a bio-impedance spectroscopy (BIS) system, we first measured complex conductivity spectra of different imaging objects including saline, agar, polyacrylamide, TX151, animal hide gelatin, banana and cucumber. Based on an analysis of how conductivity and permittivity affect measured complex boundary voltages, we suggested a new complex version of a multi-frequency time-difference image reconstruction algorithm. Imaging experiments were conducted to produce time-difference images of the objects at multiple frequencies using the proposed algorithm. Images of a conductor (stainless steel) and an insulator (acrylic plastic) were used to set a common scale bar to display all images. Comparing reconstructed time-difference images at multiple frequencies with measured complex conductivity spectra, we found that they showed an overall similarity in terms of changes in complex conductivity values with respect to frequency. However, primarily due to the limitation of the difference imaging algorithm, we suggest that multi-frequency time-difference images must be interpreted in terms of relative contrast changes with respect to frequency. We propose further imaging studies using biological tissues of known complex conductivity spectra and using human subjects to find clinical applications of the mfEIT system.
Oh
Tong In
TI
Department of Medical Physics and Bioengineering, University College London, UK.
Koo
Hwan
H
Lee
Kyung Heon
KH
Kim
Sang Min
SM
Lee
Jeehyun
J
Kim
Sung Wan
SW
Seo
Jin Keun
JK
Woo
Eung Je
EJ
D000362:Agar; D000465:Algorithms; D002138:Calibration; D003952:Diagnostic Imaging / Q000295:instrumentation* / Q000379:methods; D004553:Electric Conductivity; D017097:Electric Impedance; D005780:Gelatin; D005782:Gels; D007091:Image Processing, Computer-Assisted / Q000379:methods / Q000706:statistics & numerical data; D015203:Reproducibility of Results; D012965:Sodium Chloride; D014054:Tomography / Q000295:instrumentation* / Q000379:methods
D005782:Gels; D012965:Sodium Chloride; D005780:Gelatin; D000362:Agar