title: Capon beamforming in medical ultrasound imaging with focused beams.
abstract: Medical ultrasound imaging is conventionally done by insonifying the imaged medium with focused beams. The backscattered echoes are beamformed using delay-and-sum operations that cannot completely eliminate the contribution of signals backscattered by structures off the imaging beam to the beamsum. It leads to images with limited resolution and contrast. This paper presents an adaptation of the Capon beamformer algorithm to ultrasound medical imaging with focused beams. The strategy is to apply data-dependent weight functions to the imaging aperture. These weights act as lateral spatial filters that filter out off-axis signals. The weights are computed for each point in the imaged medium, from the statistical analysis of the signals backscattered by that point to the different elements of the imaging probe when insonifying it with different focused beams. Phantom and in vivo images are presented to illustrate the benefits of the Capon algorithm over the conventional delay and-sum approach. On heart sector images, the clutter in the heart chambers is decreased. The endocardium border is better defined. On abdominal linear array images, significant contrast and resolution enhancement are observed.
authors.affiliation: Philips Research North America, Ultrasound Imaging and Therapy, Briarcliff Manor, NY, USA. francois.vignon@philips.com
authors.forename: Francois
authors.initials: F
authors.lastname: Vignon
authors.forename: Michael R
authors.initials: MR
authors.lastname: Burcher
mesh_terms: D000465:Algorithms; D007089:Image Enhancement / Q000379:methods*; D007090:Image Interpretation, Computer-Assisted / Q000379:methods*; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity; D012815:Signal Processing, Computer-Assisted; D014463:Ultrasonography / Q000295:instrumentation / Q000379:methods*