Establishment of replacement batches for heparin low-molecular-mass for calibration CRS, and the International Standard Low Molecular Weight Heparin for Calibration.
An international collaborative study involving fourteen laboratories has taken place, organised by the European Directorate for the Quality of Medicines & HealthCare (EDQM) with National Institute for Biological Standards & Control (NIBSC) (in its capacity as a World Health Organisation (WHO) Laboratory for Biological Standardisation) to provide supporting data for the establishment of replacement batches of Heparin Low-Molecular-Mass (LMM) for Calibration Chemical Reference Substance (CRS), and of the International Reference Reagent (IRR) Low Molecular Weight Heparin for Molecular Weight Calibration. A batch of low-molecular-mass heparin was donated to the organisers and candidate preparations of freeze-dried heparin were produced at NIBSC and EDQM. The establishment study was organised in two phases: a prequalification (phase 1, performed in 3 laboratories in 2005) followed by an international collaborative study (phase 2). In phase 2, started in March 2006, molecular mass parameters were determined for seven different LMM heparin samples using the current CRS batch and two batches of candidate replacement material with a defined number average relative molecular mass (Mn) of 3,700, determined in phase 1. The values calculated using the candidates as standard were systematically different from values calculated using the current batch with its assigned number-average molecular mass (Mna) of 3,700. Using raw data supplied by participants, molecular mass parameters were recalculated using the candidates as standard with values for Mna of 3,800 and 3,900. Values for these parameters agreed more closely with those calculated using the current batch supporting the fact that the candidates, though similar to batch 1 in view of the production processes used, differ slightly in terms of molecular mass distribution. Therefore establishment of the candidates was recommended with an assigned Mna value of 3,800 that is both consistent with phase 1 results and guarantees continuity with the current CRS batch. In phase 2, participants also determined molecular weight parameters for the seven different LMM heparin samples using both the 1st IRR (90/686) and its Broad Standard Table and the candidate World Health Organization (WHO) 2nd International Standard (05/112) (2nd IS) using a Broad Standard Table established in phase 1. Mean molecular weights calculated using 2nd IS were slightly higher than with 1st IRR, and participants in the study indicated that this systematic difference precluded establishment of 2nd IS with the table supplied. A replacement Broad Standard Table has been devised on the basis of the central recalculations of raw data supplied by participants; this table gives improved agreement between values derived using the 1st IRR and the candidate 2nd IS. On the basis of this study a recommendation was made for the establishment of 2nd IS and its proposed Broad Standard Table as a replacement for the 1st International Reference Reagent Low Molecular Weight Heparin for Molecular Weight Calibration. Unlike the 1st IRR however, the candidate material 2nd IS is not suitable for use with the method of Nielsen. The candidate materials were established as heparin low-molecular-mass for calibration batches 2 and 3 by the Ph. Eur. Commission in March 2007 and as 2nd IS low-molecular-weight heparin for molecular weight calibration (05/112) by the Expert Committee on Biological Standardization in November 2007.
Mulloy
B
B
National Institute for Biological Standards and Control, South Mimms, Potters Bar, EB6 3QG Herts, United Kingdom.
Heath
A
A
Behr-Gross
M-E
ME
D000925:Anticoagulants / Q000032:analysis*; D002138:Calibration; D006495:Heparin, Low-Molecular-Weight / Q000032:analysis*; D007391:International Cooperation; D008970:Molecular Weight; D012015:Reference Standards
D000925:Anticoagulants; D006495:Heparin, Low-Molecular-Weight