The anatomic study on replacement of artificial atlanto-odontoid joint through transoral approach.
In order to provide anatomical basis for transoral approach (TOA) in dealing with the ventro lesions of craniocervical junction, and the design and application of artificial atlanto-odontoid joint, microsurgical dissecting was performed on 8 fresh craniocervical specimens layer by layer through transoropharyngeal approach. The stratification of posterior pharyngeal wall, course of vertebral artery, adjacent relationship of atlas and axis and correlative anatomical parameters of replacement of artificial atlanto-odontoid joint were observed. Besides, 32 sets of atlanto-axial joint in adults' fresh bony specimens were measured with a digital caliper and a goniometer, including the width of bony window of anterior arch of atlas, the width of bony window of axis vertebra, the distance between superior and inferior two atlas screw inserting points, the distance between two axis screw inserting points etc. It was found that the width of atlas and axis which could be exposed were 40.2+/-3.5 mm and 39.3+/-3.7 mm respectively. The width and height of posterior pharyngeal wall which could be exposed were 40.1+/-5.2 mm and 50.2+/-4.6 mm respectively. The distance between superior and inferior two atlas screw inserting points was 28.0+/-2.9 mm and 24.0+/-3.5 mm respectively, and the distance of bilateral axis screw inserting points was 18.0+/-1.2 mm. The operative exposure position through TOA ranged from inferior part of the clivus to the superior part of the C3 vertebral body. Posterior pharyngeal wall consisted of 5 layers and two interspaces: mucosa, submucosa, superficial muscular layer, anterior fascia of vertebrae, anterior muscular layer of vertebrae and posterior interspace of pharynx, anterior interspace of vertebrae. This study revealed that it had the advantages of short operative distance, good exposure and sufficient decompression in dealing with the ventro lesions from the upper cervical to the lower clivus through the TOA. The replacement of artificial atlanto-odontoid joint is suitable and feasible. The design of artificial atlanto-odontoid joint should be based on the above data.
Hu
Yong
Y
Department of Spinal Surgery, Ningbo Sixth Hospital, Ningbo 315040, China. huyong610@163.com
Yang
Shuhua
S
Xie
Hui
H
He
Xianfeng
X
Xu
Rongming
R
Ma
Weihu
W
Feng
Jianxiang
J
Chen
Qiu
Q
D001268:Atlanto-Axial Joint / Q000033:anatomy & histology* / Q000601:surgery*; D001860:Bone Plates; D001863:Bone Screws; D002102:Cadaver; D002574:Cervical Vertebrae / Q000033:anatomy & histology* / Q000601:surgery; D004867:Equipment Design; D006801:Humans; D016268:Internal Fixators; D007595:Joint Prosthesis; D008953:Models, Anatomic; D009809:Odontoid Process / Q000601:surgery*; D011474:Prosthesis Design