title: Computer simulation of the gap-tripole ion trap with linear injection, 3D ion accumulation, and versatile packet ejection.
abstract: The behavior of a completely new ion trap is shown with SIMION 7.0 simulations. The simulated trap, which was a mix of a linear and a 3D trap, was made by axially setting two ion guides with a gap between them. Each guide consisted of three rods with three symmetrically delayed radio frequency (rf) voltages (tripole). The "injected" ions were linearly contained by pulsed potentials on the entrance and exit plates. Then the three-dimensional (3D) rf field in the gap, which was created by the tripole special rod arrangement, could trap the ions when the translational energy was dampened by collisions with low-pressure nitrogen. Because the injected ions were trapped in the small gap, the trapping cycle could be repeated many times before ion ejection, so a high concentrated ion cloud could be obtained. This trapping and accumulation methodology is not possible in most conventional multipole linear traps with even number of poles. Compared with quadrupole linear trap at the same rf amplitude, tripole lost more ions due to strong charge repulsion in the ion cloud. However, tripole could catch up the ions at higher voltage. Radial and axial mass-independent ejection of the ions localized in the tripole gap was very simple, compared with conventional linear ion traps that need extra and complicated electrodes for effective axial ejection.
authors.affiliation: Analytical Molecular Medicine and Devices Laboratory, Faculty of Frontier Medical Sciences, Graduate School of Hiroshima University, Minami-ku, Hiroshima, Japan.
authors.forename: Gary A
authors.initials: GA
authors.lastname: Salazar
authors.forename: Tsutomu
authors.initials: T
authors.lastname: Masujima
mesh_terms: D003198:Computer Simulation; D017076:Computer-Aided Design; D004574:Electromagnetic Fields; D004867:Equipment Design; D013058:Mass Spectrometry / Q000295:instrumentation*; D013816:Thermodynamics