[The study of aluminium diffuser calibration in the UV].
A bi-directional reflectance distribution function (BRDF) measurement setup in the ultraviolet spectral range was established. The BRDF of the aluminium diffusers at a given orientation was measured. The relative accuracy of the BRDF measurement is better than 2.5%. The hemispheric reflectance of the aluminium diffusers was measured in the wavelength range from 250 to 650 nm. It increases with the wavelength, and changes about 6% from 300 to 360 nm. It decreases with the time. Since the diffuser was made (about one year ago), from 250 to 300 nm, the peak decrease in the hemispheric reflectance can reach 2.6%, and the average decrease is 1.5%. From 300 to 360 nm, it has an average decrease of 0.9% decrease, and 0.8% when wavelength is longer than 360 nm.
Li
Cong
C
Lab for Space Environment Exploration, Center for Space Science and Applied Research, Chinese Academy of Sciences, Beijing 100080, China. licongnc@hotmail.com
Wang
Yong-Mei
YM
Zhang
Zhong-Mou
ZM